Varying negative work assistance at the ankle with a soft exosuit during loaded walking
© The Author(s). 2017
Received: 21 October 2016
Accepted: 1 June 2017
Published: 26 June 2017
Only very recently, studies have shown that it is possible to reduce the metabolic rate of unloaded and loaded walking using robotic ankle exoskeletons. Some studies obtained this result by means of high positive work assistance while others combined negative and positive work assistance. There is no consensus about the isolated contribution of negative work assistance. Therefore, the aim of the present study is to examine the effect of varying negative work assistance at the ankle joint while maintaining a fixed level of positive work assistance with a multi-articular soft exosuit.
We tested eight participants during walking at 1.5 ms−1 with a 23-kg backpack. Participants wore a version of the exosuit that assisted plantarflexion via Bowden cables tethered to an off-board actuation platform. In four active conditions we provided different rates of exosuit bilateral ankle negative work assistance ranging from 0.015 to 0.037 W kg−1 and a fixed rate of positive work assistance of 0.19 W kg−1.
All active conditions significantly reduced metabolic rate by 11 to 15% compared to a reference condition, where the participants wore the exosuit but no assistance was provided. We found no significant effect of negative work assistance. However, there was a trend (p = .08) toward greater reduction in metabolic rate with increasing negative work assistance, which could be explained by observed reductions in biological ankle and hip joint power and moment.
The non-significant trend of increasing negative work assistance with increasing reductions in metabolic rate motivates the value in further studies on the relative effects of negative and positive work assistance. There may be benefit in varying negative work over a greater range or in isolation from positive work assistance.
Through regular training, individuals in certain specialized professions are able to walk fast with heavy loads. For example, soldiers are able to walk with loads ranging from 30 to 100% their body weight . Nevertheless, overloading dramatically increases the metabolic rate of walking  which in turn causes earlier onset of fatigue and increased injury risk .
Different labs have developed robotic exoskeletons intended to unburden individuals carrying load. While early studies with rigid full-leg exoskeletons demonstrated the ability to unload a backpack load from a wearer , these systems caused an increase in metabolic effort compared to normal walking [5, 6]. However, over the past years, a number of labs successfully developed early prototype devices that achieved reductions in metabolic rate of unloaded [7–9] and loaded [10, 11] walking, and increase in maximal load-carrying performance  in controlled treadmill experiments. One explanation as to why reductions in metabolic rate were achieved only recently could be that earlier exoskeletons were too heavy [10, 13], with rigid structures over the length of the whole leg that caused wearers to deviate significantly from their natural gait. Recent single joint designs [7, 8, 10, 14–17] weigh less and provide targeted assistance at the wearer’s joints, rather than transferring load directly to the ground. Apart from the improved designs, an additional factor that contributed to the recent successes of these assistive devices was knowledge gained from parameter sweep studies that explored the effects of different actuation parameters in a systematic way [7, 8, 18–20].
Two devices that produced metabolic rate reduction versus normal walking provided high positive work assistance at the ankle joint [7, 9, 10]. Another device from Collins et al., however, produced metabolic reduction by means of an intermediate level of moment assistance during both the period of negative and positive ankle work using an unpowered ankle exoskeleton with a clutchable spring .
Following the path of wearable robots that target specific joints, our group has developed lightweight and soft wearable robots (“exosuits”), that assist the ankle and hip joint during (fast) loaded walking [11, 18, 21, 22], mostly for applications in first responders and soldiers for whom high speed walking is a job requirement. Exosuits use textiles to interface with the body and apply joint moments via Bowden cables pulling over the outside of the body in parallel with the muscles, using the biological skeleton to support compressive loads. Advantages of exosuits are that they have very low distal inertia and do not require precise alignment with biological joints. Recently, we achieved metabolic reductions with an autonomous version of our exosuit compared to wearing the exosuit powered-off with the equivalent mass of the actuation hardware and power source removed . This version of our exosuit provided a combination of mostly positive work assistance and low negative work assistance. In contrast to earlier mentioned single-joint devices this exosuit assisted plantarflexion, hip extension and hip flexion.
Studies about devices that only assist the positive work phase of the ankle joint often motivate this actuation design choice based on the knowledge that concentric muscle work is metabolically more expensive , and that half of the sagittal plane lower limb positive mechanical joint work is required at the ankle . In the study with the negative work phase assistance, only very low positive work assistance was provided . The authors hypothesized that the reduction in metabolic rate was not due to a reduction in joint or muscle work, but rather to a reduction in isometric muscle force, while the Achilles tendon is being elongated. This claim is based on a simulation model from Umberger  that predicts that most of the metabolic energy consumption from the plantarflexors occurs during single stance. However, there are also recent indications that providing assistance during single stance may shift the plantarflexor muscle fascicle length toward unfavorable contractile conditions, which may disrupt the push-off phase that follows thereafter .
Studies that promote positive joint work assistance, and others that promote negative joint work assistance at the ankle in reducing metabolic cost are not mutually exclusive. One difficulty in reconciling these two conclusions comes from the fact that no study to date has attempted to vary the contribution of the negative work phase of the ankle joint. Therefore, the aim of the present study was to investigate the effect of negative work assistance at the ankle by varying the rate of negative work assistance de-coupled from the rate of positive work assistance. Going forward with our design approach, we chose to deliver the assistance with a version of our soft exosuit that assisted mostly the ankle. We also chose to test the effects during fast loaded walking in order to provide representative data on how to assist the ankle in first-responders or soldiers. The selection of loaded walking could lead to additional benefits for assisting negative work since biological negative joint work at the ankle is increased during loaded walking .
While no other study has varied negative joint work at the ankle over multiple conditions, one study from Jackson and Collins  contained one condition with only negative work assistance at the ankle, and compared this to a powered-off condition. Based on their dataset, we can hypothesize that varying negative work assistance will lead to an increase in metabolic cost by 8.2 J per J negative work assistance.1 Alternatively, based on the energy cost of eccentric muscle work reported by Margaria  we can hypothesize that assisting negative work at the ankle can potentially reduce metabolic cost by 1.2 J per J negative work assistance if negative work assistance would lead to an equal amount of reduction in biological work and if the efficiency of this saved biological ankle work were the same as the whole-body efficiency of downhill walking that was found by Margaria. Therefore we hypothesize that varying negative work assistance at the ankle will lead to reductions in metabolic rate between a 1.2 and 8.2 J per J negative work assistance.
Additional file 1: Movie Participant walking with the exosuit in the Powered-Off, Minimal and High negative work rate condition. (M4 V 42521 kb)
Off-board actuation platform
Because the goal of this study was to examine the relative effect of varying negative work assistance at the ankle joint while maintaining a fixed level of positive work assistance with a multi-articular soft exosuit, we used an off-board actuation platform similar to the one described in [20, 29–31], but with an alternative electromechanical design (Fig. 1). It consisted of two Maxon® EC motors (EC30, Maxon Motor AG, Sachseln, Switzerland; 200 W), and each motor drove a 3-cm radius pulley through a 55:1 planetary gearbox. These actuators were similar to the ones used in a fully autonomous exosuit  so the actuation profiles that we tested were within a range of profiles that are realistically achievable with an autonomous exosuit. We used Bowden cables to transmit the forces from the actuator to the exosuit. The end of the Bowden cable sheaths attached to the back of the calf wrap, while the inner cable extends further to the back of the heels. When the motor rotates inward, this generates forces between these two cable attachment points and also along the entire length of the load path of the multi-articular straps. The control board was a modified version of Arduino Due; it was composed of two servomotor drivers (Gold Twitter, Elmo Motion Control Ltd., Nashua, NH, USA) and two ARM-based processors (Cortex-M3, Atmel Corporation, San Jose, CA, USA), which controlled the two actuators respectively, together with CAN communication among all subsystems.
We measured real-time data from the wearer and the exosuit by means of a load cell (LTH300, Futek Advanced Sensor Technology Inc., Irvine, CA, USA) and two single-axis gyroscopes (LY3100ALH, STMicroelectronics N.V., Geneva, Switzerland) per leg. We attached two gyroscopes on the boot: one at the dorsal side of the midfoot and the other at the anterior side of the shank. We aligned the axes perpendicular to the flexion-extension axis and calculated ankle angular velocity by subtracting the angular velocities measured by both sensors.
We used a control algorithm based on an iterative force-based position controller that we described in more detail in . The controller detected the beginning of the stance phase based on initial plantarflexion from the foot gyroscope. To provide negative work assistance the motor held the cable at a fixed position and passively generated force due to the effective human-suit stiffness [32, 33] as the wearer dorsiflexed. Since in normal walking the biological ankle moment is in the plantarflexion direction during the entire part of the stance phase after initial forefoot contact we know that at the ankle angular velocity zero-crossing point, the biological ankle power becomes positive. Thus, detection of ankle angular velocity zero-crossing allowed us to determine the timing for pulling on the cable to assist the positive work phase. About halfway through the push-off phase, the cable force automatically drops because the ankle joint further plantarflexes. When the controller detected this, the motor spooled the cable to prevent resistance during the swing phase.
The controller calculated exosuit power for each ankle in real time based on the product of ankle angular velocity and exosuit ankle moment (based on the load cell force and moment arm). Next, the controller calculated positive and negative exosuit work rates for each ankle respectively by integrating the positive and negative power portions and dividing them by the stride time. The controller compared measured negative and positive work rates to desired values and iteratively adjusted the pretension and active force for each ankle on a stride-by-stride basis similar to [29, 34, 35]. At the beginning of each walking condition the controller required about one minute to reach the desired rate of negative work assistance.
It should be noted that due to the multi-articular strap the exosuit also passively applied positive and negative work at the hip joint. The amount of work at the hip resulted passively from the forces applied at the ankle and the amount by which the straps were tightened. It was not directly controlled since we did not have separate actuation at the hip.
Eight healthy male participants (26.3 ± 4.7 y, 79.9 ± 9.5 kg, 1.78 ± 0.06 m) who were experienced load carriers or endurance athletes participated in this study. The Harvard Medical School Committee on Human Studies approved the study and the participants provided written informed consent. The participants whose images appear in the manuscript have provided written consent for the publication of their images according to the policies of the Journal of NeuroEngineering and Rehabilitation.
Participants attended two experimental sessions, a training and a testing session that were 2 to 7 days apart. Appropriate sizes of the exosuit components were fitted during training. After donning the system, participants walked on a treadmill at 1.5 m s−1 while carrying a load of 23-kg and experienced all of the different actuation conditions for 8 min each, for a total of 32 min of training . During the testing session participants completed the same protocol as in the training plus an additional 8-min baseline condition. In this condition, called Powered-Off, the participants wore the exosuit and the backpack but the cables were slack such that they had close to zero tension so that no moments were applied to the joints. We randomized the order of the conditions.
During the testing session we measured respiratory gases by means of a computerized O2-CO2 analyzer and flow meter (K4b2, COSMED, Rome, Italy) during a 4-min standing trial and during the walking conditions. We recorded 3D kinematics using motion capture (Vicon, Oxford Metrics, Oxford, UK). We attached reflective markers to the boots, shank, thigh, pelvis, trunk and Bowden cables attachments. We acquired ground reaction force data from split-belt instrumented treadmill (Bertec Corporation, Columbus, OH, USA).
We calculated metabolic rate by means of the Brockway equation  and averaged the last 2 min of each trial. We averaged only the last two minutes in order to allow sufficient time for the exosuit actuation to stabilize and for metabolic rate to reach steady state. We filtered raw kinematic and ground reaction force data using a zero-lag, low-pass, fourth-order Butterworth filter with a cut-off frequency between 8 Hz and 15 Hz, as determined by preliminary residual analysis. We calculated sagittal plane joint kinetics and kinematics using Visual3D (C-motion Inc., Rockville, MD, USA). We calculated the exosuit moments based on a moment arm determined from markers on the exosuit load path and joint axes. We calculated biological joint kinetics by subtracting the exosuit joint kinetics from the total external joint kinetics [7, 8, 11, 15, 16, 19, 20]. We calculated center-of-mass displacement by dividing total ground reaction force by body mass and integrating over time . We calculated center-of-mass power for each leg by taking the dot product of center-of-mass velocity and the individual leg ground reaction force . We determined heel contact and toe off times based on ground reaction forces and kinematics using the automatic gait detection algorithm from Visual3D . We averaged kinematics and kinetics of 10 strides of the last minute of each walking condition and plotted the results versus stride percentage. After each condition we gave the participants a questionnaire in which we asked them to score the perceived assistance and perceived comfort on a VAS scale similar to . We excluded data from one participant because we noticed that the waist belt slipped during testing, which resulted in the system malfunctioning and large standard deviations in the physiological data. In addition, for two participants we were unable to provide the desired negative work rate in High, due to the participants’ specific gait kinematics and the performance limitations of the off-board actuation platform. For these two participants we only collected and analyzed data from three active conditions.
We used mixed-model, two-factor ANOVA with participants as random-effect to test the effect of of negative exosuit ankle work rate. To evaluate differences between the conditions we performed repeated-measures ANOVA. If the p-value was equal to or lower than 0.05, we conducted pairwise comparisons between Powered-Off and the active conditions with paired t-tests with Šídák-Holm correction for multiple testing. In order to evaluate where significant effects were situated in time with respect to the actuation period, we performed statistics for all frames of the stride cycle and visually represented results in colorbars above time series charts similar to [41, 42]. We also conducted the same statistical tests on relevant single-value metrics (e.g. metabolic rate). We reported all inter-subject variabilities as standard errors of the means.
Metabolic rates of walking minus resting in Minimal, Low, Medium and High respectively were 5.21 ± 0.20, 5.10 ± 0.16, 5.20 ± 0.21 and 5.00 ± 0.24 W kg−1. Metabolic rate of walking minus resting in Powered-Off was 5.84 ± 0.20 W kg−1. In all the active conditions metabolic rate was significantly lower than in Powered-Off (p ≤ 5 · 10−4, paired t-tests). Reductions in metabolic rate compared to Powered-Off in Minimal, Low, Medium and High respectively were 0.62 ± 0.09, 0.73 ± 0.14, 0.63 ± 0.09, 0.88 ± 0.17 W kg−1. Percent reductions Compared to Powered-off were 11 ± 1, 12 ± 2, 11 ± 2 and 15 ± 3% (Fig. 3b).
There was no significant effect of negative exosuit ankle work rate on reduction in metabolic rate versus Powered-Off but there was a trend toward significance (p = 0.083). The average efficiency of negative exosuit ankle work assistance is visualized by the slope of this trend in Fig. 3b which amounts to 8.8 W reduction in metabolic rate per W negative work rate from both exosuit ankles. The average efficiency of positive exosuit ankle work assistance can be derived from the intercept value of this trend with a vertical line through zero negative work rate on the horizontal axis, divided by the positive exosuit ankle work rate. This efficiency of positive work assistance was 2.7 W reduction in metabolic rate per W total positive exosuit ankle work rate.
In Minimal there was less maximal knee flexion during early stance than in the Powered-Off (p = 0.030, Fig. 6b). Increasing negative exosuit ankle work rate led to reduced maximum hip extension (p = 0.015, Fig. 6c).
Biological joint kinetics
Estimated biological hip extension and flexion moment decreased with increasing negative exosuit ankle work rate during early stance and the beginning of push-off (p ≤ 0.05 from 24 to 54% of the stride cycle, Fig. 7c). In all active conditions, estimated biological hip flexion moment was reduced compared to Powered-Off during the push-off phase of the ipsilateral leg (p ≤ 0.05 from 13 to 18% of the stride period), as well as during the push-off phase of the contralateral leg (p ≤ 0.05 from 54 to 64% of the stride period).
The estimated biological hip power decreased with increasing negative exosuit ankle work rate, from 23 to 50% of the stride cycle (p ≤ 0.05). Both negative and positive estimated biological hip power during the push-off phase showed significant reductions compared to Powered-Off in all the active conditions (p ≤ 0.05, Fig. 8c).
Total body kinetics
All the active conditions significantly reduced metabolic rate compared to Powered-Off by 11 to 15%. There was a trend showing increasing reduction in metabolic rate with increasing negative work assistance (Fig. 3b). This trend followed an average slope of 8.8 J metabolic reduction per J negative work assistance. This slope was different from the expected range based on ankle work which was between slope of 8 J metabolic increase per J negative work  and 1.2 J metabolic reduction per J negative work . However, the slope of the trend in our study was not significant.
There are different possible reasons as to why there was no significant effect of negative work assistance on metabolic rate. The negative work assistance provided in High was only about half the negative work in the optimal stiffness condition in the study from Collins et al. . In this study we were limited in our ability to provide higher negative work due to the fact that the controller used maintained the ankle cable at a fixed length during the pretension phase. Additionally, due to the fast walking velocity of our study (1.5 m s−1), the duration of the dorsiflexion phase was relatively short, which limits the percentage of the gait cycle where negative work can be applied [43, 44]. Even if the negative work assistance would have reduced metabolic rate with the same efficiency as eccentric contractions in biological muscles , the range of negative work assistance that we applied would only have led to a change in metabolic rate of 4%—close to the observed average difference between Minimal and High. Similar to the study from Collins et al.  we also found that peak dorsiflexion decreased with increasing negative exosuit ankle work rate. This could potentially put the plantar flexors in disadvantageous contractile conditions for push off by reducing Achilles tendon stretch [8, 45].
Conversely, there are different possible explanations as to why we still found a small trend toward higher reduction in metabolic rate with increasing negative work assistance. We found significant effects of negative exosuit ankle work resulting increases in estimated exosuit hip power and moment and reductions in estimated biological hip power during the negative work assistance phase. These are likely the effects of the force transfer from the multi-articular straps or indirect effects of plantarflexion assistance [17, 45] or a combination of both. We also found reductions in biological ankle peak power and moment with increasing negative work assistance probably due to the fact that the participants and the controller generated higher peak force in increasing negative work assistance conditions. Finally, we found higher reductions in the center-of-mass rebound work with increasing negative work assistance.
One limitation for interpreting the effect of negative work assistance at the ankle is that we did not keep the hip assistance constant. While we were able to rigorously control the exosuit ankle work and to maintain the positive portion constant, this also resulted in changes in the exosuit hip power, though to a lesser extent than in the ankle actuation. In this version of the exosuit, the use of the multi-articular strap that couples ankle actuation to hip actuation was a necessary compromise in order to keep the calf wrap from slipping. This is a common challenge of parameter sweeps with exoskeletons. As control schemes improve, we are now able to very rigorously manipulate certain parameters of interest and keep others constant. However, it will never be possible to keep all parameters constant, even with single-joint exoskeletons, simply because certain parameters inherently vary with the parameter of interest. Moreover, we only measured the multi-articular strap force distribution in tests that were separate from the biomechanical and physiological evaluation reported here so the reported exosuit hip moments and powers are only estimations. Further, we calculated the changes in metabolic rate compared to a Powered-Off condition and not compared to a no-suit condition; thus, the effect of wearing the exosuit components is evaluated in the results. This was done to simplify the protocol and reduce the time between relative metabolic comparisons such that subjects did not have to stop and doff the exosuit between conditions. It is currently unknown how such a comparison would change the results, but due to the lightweight nature of the exosuit components (about one kg total distributed on both legs, Additional file 5), we expect that the difference would be small. For this reason, however, the augmentation factor proposed by Mooney et al. [9, 10] was not calculated in this study due to the lack of a strict no-suit condition.
The derived efficiency of positive work assistance of 2.7 J metabolic reduction per J mechanical work falls within the range of ratios reported in rigid ankle exoskeletons studies (1.6 in , 2.7 in , 3.9 in , 4.7 in ). Possible explanations for the benefits of positive work assistance can be found in differences in biomechanics versus Powered-Off that occurred during the positive work assistance phase. For example, we found reductions in biological ankle moment during push-off with respect to Powered-Off. We also found reductions in biological hip flexion power and biological hip extension and flexion moment during this phase.
The observed benefits of positive work assistance together with the absence of a significant metabolic benefit of negative work assistance suggest that for autonomous applications of this exosuit, focusing on ankle positive power assistance should be considered and there may be small additional benefits, and no major biomechanical disadvantages of also providing assistance during the ankle negative power phase. Additional considerations could also be on the actuation requirements to realize the intended assistance and how this would affect system and battery mass. It is likely that these results are specific to the architecture of the exosuit used here and different results could be found when assisting different joints with other exosuits architectures or rigid exoskeletons or with different carried loads. It is also possible that a different result would be found if the negative and positive work assistance would be applied with different force profile shapes. More research is needed to provide a definitive answer on when negative work assistance is beneficial or detrimental.
This study describes how to provide both negative and positive work assistance at the ankle with a soft exosuit during loaded walking by providing a low pretension force at the heel while the ankle dorsiflexes and actively pulling as soon as the ankle starts plantarflexing. All the active conditions significantly reduced metabolic rate on average by 11 to 15% versus Powered-Off. We found a trend toward more reduction in metabolic rate with increasing negative work assistance but this was not significant. However, we did see significant benefits of negative work assistance on biological ankle and hip joint kinetics which could explain the trend toward significant metabolic reduction. Positive work assistance led to high reductions in metabolic rate relative to the provided positive work. The non-significant trend of increasing negative work assistance with increasing reductions in metabolic rate motivates the value in further studies on the relative effects of negative and positive work assistance. In particular, there may be benefit in investigating varying negative work over a greater range or in varying negative work assistance without providing positive work assistance.
We calculated the ratio of change in metabolic rate per unit negative work rate assistance based on the dataset from Jackson and Collins  that was made available for download in . From their dataset, we calculated the increase in metabolic cost in the condition where only negative work was provided by subtracting the metabolic rate in the powered-off condition from the condition in which only negative work was provided. Then, we calculated the rate of negative work that was provided in that condition by integrating the exoskeleton ankle joint power over time and dividing by stride time. Finally, we divided the increase in metabolic rate by the negative exoskeleton ankle work rate to obtain an individual ratio for every participant. The mean of this ratio is 8.2 J increase in metabolic rate per J negative work.
The authors would like to thank Adam Couture, Alan Asbeck, Diana Wagner for their contribution to this work.
This material is based upon the work supported by the Defense Advanced Research Projects Agency (DARPA), Warrior Web Program (Contract No. W911NF-14-C-0051), the National Science Foundation (Grant No. CNS-1446464), the National Science Foundation Graduate Research Fellowship Program (Grant No. DGE1144152), the Samsung Scholarship. This work was also partially funded by the Wyss Institute and the John A. Paulson School of Engineering and Applied Sciences at Harvard University.
Availability of data and materials
All the data relevant to this study are presented within the manuscript.
CW, FAP, IG, KGH, PM, SC, SL designed the experiments, IG, SC, SL designed the control algorithm. FS, PM, SC, SL performed the biomechanics experiments. CS, FS, KGH, PM, SC, SL analyzed and interpreted the data. CS, CW, FS, FAP, IG, KGH, PM, SC, SL wrote the manuscript. All authors read and approved the final manuscript.
Patents have been filed with the U.S. Patent Office describing the exosuit components documented in this manuscript. S.L., I.G. and C.J.W were authors of those patents. There is no other competing interest to declare. Harvard University has entered into a license and collaboration agreement with ReWalk Robotics.
Consent for publication
Ethics approval and consent to participate
All participants provided written informed consent prior to participating in the study. The study was approved by the Harvard Medical School Committee on Human Studies. The participants whose images appear in the manuscript have provided written consent for the publication of their images according to the policies of the Journal of NeuroEngineering and Rehabilitation.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
Open AccessThis article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.
- Caron RR, Lewis CL, Saltzman E, Wagenaar RC, Holt KG. Musculoskeletal stiffness changes linearly in response to increasing load during walking gait. J Biomech. 2015;48:1165–71.View ArticlePubMedGoogle Scholar
- Bastien GJ, Willems PA, Schepens B, Heglund NC. Effect of load and speed on the energetic cost of human walking. Eur J Appl Physiol. 2005;94:76–83.View ArticlePubMedGoogle Scholar
- Knapik J, Reynolds K, Harman E. Soldier load carriage: historical, physiological, biomechanical, and medical aspects. Mil Med. 2004;169:45–56.View ArticlePubMedGoogle Scholar
- Walsh CJ, Paluska D, Pasch K, Grand W, Valiente A, Herr H. Development of a lightweight, underactuated exoskeleton for load-carrying augmentation. Orlando: Int. Conf. Robot. Autom.; 2006. p. 3485–3491. http://0-ieeexplore.ieee.org.brum.beds.ac.uk/abstract/document/1642234/.
- Gregorczyk KN, Hasselquist L, Schiffman JM, Bensel CK, Obusek JP, Gutekunst DJ. Effects of a lower-body exoskeleton device on metabolic cost and gait biomechanics during load carriage. Ergonomics. 2006;53:1263–75.View ArticleGoogle Scholar
- James Walsh C, Endo K, Herr H. A quasi-passive leg exoskeleton for load carrying augmentation. Int J Humanoid Robot. 2007;4:487–506.View ArticleGoogle Scholar
- Malcolm P, Derave W, Galle S, De Clercq D. A simple exoskeleton that assists plantarflexion can reduce the metabolic cost of human walking. PLoS One. 2013;8:1–7.Google Scholar
- Collins SH, Wiggin MB, Sawicki GS. Reducing the energy cost of human walking using an unpowered exoskeleton. Nature. 2015;522:212–5.View ArticlePubMedPubMed CentralGoogle Scholar
- Mooney LM, Rouse EJ, Herr HM. Autonomous exoskeleton reduces metabolic cost of human walking. J. Neuroeng. Rehabil. 2014;11:1–5.View ArticleGoogle Scholar
- Mooney LM, Rouse EJ, Herr HM. Autonomous exoskeleton reduces metabolic cost of human walking during load carriage. J. Neuroeng. Rehabil. 2014;11:80.View ArticlePubMedPubMed CentralGoogle Scholar
- Panizzolo FA, Galiana I, Asbeck AT, Siviy C, Schmidt K, Holt KG, et al. A biologically-inspired multi-joint soft exosuit that can reduce the energy cost of loaded walking. J Neuroeng Rehabil. 2016;13:43.View ArticlePubMedPubMed CentralGoogle Scholar
- Galle S, Malcolm P, Derave W, De Clercq D. Enhancing performance during inclined loaded walking with a powered ankle-foot exoskeleton. Eur J Appl Physiol. 2014;114:2341–51.View ArticlePubMedGoogle Scholar
- Herr H. Exoskeletons and orthoses: classification, design challenges and future directions. J. Neuroeng. Rehabil. 2009;6:21.View ArticlePubMedPubMed CentralGoogle Scholar
- Norris JA, Granata KP, Mitros MR, Byrne EM, Marsh AP. Effect of augmented plantarflexion power on preferred walking speed and economy in young and older adults. Gait Posture. 2007;25:620–7.View ArticlePubMedGoogle Scholar
- Jackson RW, Collins SH. An experimental comparison of the relative benefits of work and torque assistance in ankle exoskeletons. J Appl Physiol. 2015;119:541–57.View ArticlePubMedGoogle Scholar
- Sawicki GS, Ferris DP. Mechanics and energetics of level walking with powered ankle exoskeletons. J Exp Biol. 2008;211:1402–13.View ArticlePubMedGoogle Scholar
- Koller JR, Jacobs DA, Ferris DP, Remy CD. Learning to walk with an adaptive gain proportional myoelectric controller for a robotic ankle exoskeleton. J. Neuroeng. Rehabil. 2015;12:97.View ArticlePubMedPubMed CentralGoogle Scholar
- Wehner M, Quinlivan B, Aubin PM, Martinez-Villalpando E, Baumann M, Stirling L, et al. A lightweight soft exosuit for gait assistance. Karlsruhe: ICRA; 2013. p. 3362–9.Google Scholar
- Galle S, Malcolm P, Collins SH, Speeckaert J, De Clercq D. Optimizing robotic exoskeletons actuation based on human neuromechanics experiments: interaction of push-off timing and work. Cambridge: AMAM; 2015. p. 1–3.Google Scholar
- Ding Y, Panizzolo FA, Siviy CJ, Malcolm P, Galiana I, Holt KG, et al. Effect of timing of hip extension assistance during loaded walking with a soft exosuit. J Neuroeng Rehabil. 2016;13:87.View ArticlePubMedPubMed CentralGoogle Scholar
- Asbeck A, De Rossi SMM, Galiana I, Ding Y, Walsh C. Stronger, smarter, softer: next-generation wearable robots. IEEE Robot Autom. 2014;21:22–33.View ArticleGoogle Scholar
- Asbeck AT, De Rossi SMM, Holt KG, Walsh CJ. A biologically inspired soft exosuit for walking assistance. Int J Robot Res. 2015;34:744–62.View ArticleGoogle Scholar
- Margaria R. Positive and negative work performances and their efficiencies in human locomotion. Int Zeitschrift fur Angew Physiol. Einschl. Arbeitsphysiologiete Physiol. Einschl. Arbeitsphysiologie. 1968;25:339–51.Google Scholar
- Winter D. Energy generation and absorption at the ankle and knee during fast, natural, and slow cadences. Clin Orthop Relat Res. 1983;175:147–54.Google Scholar
- Umberger BR. Stance and swing phase costs in human walking. J R Soc Interface. 2010;7:1329–40.View ArticlePubMedPubMed CentralGoogle Scholar
- Dick, T.J.M., Nuckols, R.W., Sawicki, G.S., 2017. Tuned or not? Ultrasound measurements of soleus fascicle dynamics during human walking with elastic ankle exoskeletons, in: American Society of Biomechanics.Google Scholar
- Huang T-WP, Kuo AD. Mechanics and energetics of load carriage during human walking. J Exp Biol. 2014;217:605–13.View ArticlePubMedPubMed CentralGoogle Scholar
- Asbeck AT, Schmidt K, Galiana I, Wagner D, Walsh CJ. Multi-joint Soft Exosuit for Gait Assistance. Seattle: ICRA; 2015. p. 6197–204.Google Scholar
- Lee S, Crea S, Malcolm P, Galiana I, Asbeck A, Walsh C. Controlling negative and positive power at the ankle with a soft exosuit. Stockholm: ICRA; 2016. p. 3509–15.Google Scholar
- Ding Y, Galiana I, Asbeck A, Quinlivan B, De Rossi SMM, Walsh CJ. Multi-joint actuation platform for lower extremity soft exosuits. Int. Conf. Robot. Autom. Hong Kong: Ieee; 2014. p. 1327–34.Google Scholar
- Ding Y, Galiana I, Asbeck A, De Rossi S, Bae J, Santos T, et al. Biomechanical and Physiological Evaluation of Multi-joint Assistance with Soft Exosuits. TNSRE. 2016;25(2):119–30. doi:10.1109/TNSRE.2016.2523250, http://0-ieeexplore.ieee.org.brum.beds.ac.uk/abstract/document/7394183/.
- Asbeck AT, De Rossi SMM, Holt KG, Walsh CJ. A biologically inspired soft exosuit for walking assistance. Int J Rob Res. 2015;34(6):744–62. https://doi.org/10.1177/0278364914562476.
- Lee S, Crea S, Galiana I, Malcolm P, Walsh CJ. Controlling negative and positive power at the ankle with a soft exosuit. Stockholm: Int. Conf. Robot. Autom; 2016.View ArticleGoogle Scholar
- Panizzolo F, Galiana I, Asbeck AT, Siviy C, Schmidt K, Holt KG, et al. A biologically-inspired multi-joint soft exosuit that can reduce the energy cost of loaded walking. J. Neuroeng. Rehabil.. Journal of NeuroEngineering and Rehabilitation; 2016 http://0-dx.doi.org.brum.beds.ac.uk/10.1186/s12984-016-0150-9
- Malcolm P, Quesada RE, Caputo JM, Collins SH. The influence of push-off timing in a robotic ankle-foot prosthesis on the energetics and mechanics of walking. J Neuroeng Rehabil. 2015;12:21.View ArticlePubMedPubMed CentralGoogle Scholar
- Galle S, Malcolm P, Derave W, De Clercq D. Adaptation to walking with an exoskeleton that assists ankle extension. Gait Posture. 2013;38:495–9.View ArticlePubMedGoogle Scholar
- Brockway JM. Derivation of formulae used to calculate energy expenditure in man. Hum Nutr Clin Nutr. 1987;41:463–71.PubMedGoogle Scholar
- Cavagna GA. Force platforms as ergometers. J Appl Physiol. 1975;39:174–9.PubMedGoogle Scholar
- Donelan JM, Kram R, Kuo AD. Simultaneous positive and negative external mechanical work in human walking. J Biomech. 2002;35:117–24.View ArticlePubMedGoogle Scholar
- Stanhope SJ, Kepple TM, McGuire DA, Roman NL. Kinematic-based technique for event time determination during gait. Med Biol Eng Comput. 1990;28:355–60.View ArticlePubMedGoogle Scholar
- Shamaei K, Member S, Member MC, Adams AA, Gregorczyk KN, Schiffman JM, et al. Effects of exoskeletal stiffness in parallel with the knee on the motion of the human body center of mass during walking. Seattle: ICRA; 2015. p. 5557–64.Google Scholar
- Voloshina AS, Ferris DP. Biomechanics and energetics of running on uneven terrain. J Exp Biol. 2013;216:711–9.View ArticleGoogle Scholar
- Hansen AH, Childress DS, Miff SC, Gard SA, Mesplay KP. The human ankle during walking: implications for design of biomimetic ankle prostheses. J Biomech. 2004;37:1467–74.View ArticlePubMedGoogle Scholar
- Farris DJ, Sawicki GS. The mechanics and energetics of human walking and running: a joint level perspective. J R Soc Interface. 2011;9(66):110–8. https://doi.org/10.1098/rsif.2011.0182.
- Farris DJ, Sawicki GS. Linking the mechanics and energetics of hopping with elastic ankle exoskeletons. J Appl Physiol. 2012;113:1862–72.View ArticlePubMedGoogle Scholar
- Jackson, R.W., Collins, S.H., 2015. An experimental comparison of the relative benefits of work and torque assistance in ankle exoskeletons, dataset http://biomechatronics.cit.cmu.edu/publications/Jackson_2015_JAP-%2D-Data.zip.