- Open Access
Haptic wearables as sensory replacement, sensory augmentation and trainer – a review
© Shull and Damian. 2015
Received: 13 January 2015
Accepted: 13 July 2015
Published: 20 July 2015
Sensory impairments decrease quality of life and can slow or hinder rehabilitation. Small, computationally powerful electronics have enabled the recent development of wearable systems aimed to improve function for individuals with sensory impairments. The purpose of this review is to synthesize current haptic wearable research for clinical applications involving sensory impairments. We define haptic wearables as untethered, ungrounded body worn devices that interact with skin directly or through clothing and can be used in natural environments outside a laboratory. Results of this review are categorized by degree of sensory impairment. Total impairment, such as in an amputee, blind, or deaf individual, involves haptics acting as sensory replacement; partial impairment, as is common in rehabilitation, involves haptics as sensory augmentation; and no impairment involves haptics as trainer. This review found that wearable haptic devices improved function for a variety of clinical applications including: rehabilitation, prosthetics, vestibular loss, osteoarthritis, vision loss and hearing loss. Future haptic wearables development should focus on clinical needs, intuitive and multimodal haptic displays, low energy demands, and biomechanical compliance for long-term usage.
Sensory impairments, including somatosensory, vision, and audition loss can result from a spectrum of injuries and diseases such as limb loss, vision loss, and stroke and have long been known to reduce quality of life and prolong rehabilitation [1, 2]. As the world population ages, the magnitude of these problems will likely increase given the susceptibility to sensory impairments in older populations . In the absence of treatments that completely restore natural sensory function, approaches focused on replacing or augmenting deficits may serve as effective alternatives.
Human skin has long been recognized as a receptor for communicating information . Skin sensations such as pressure, vibration, and stretch can convey tactile messages that are carried to the brain via afferent nerves [5, 6]. For example, tactile feedback can be used to encode pressure and vibration measurements from a prosthesis to the skin of a user . To train human movement, kinematics can be measured in real time and compared with predefined desired kinematics, and tactile feedback amplitude or frequency can then be modulated proportionally to error signals to alert users of desired changes [8–10]. Similarly, tactile feedback has been used to train repetitive movements such as swimming or gait [11–13] in which case feedback is initiated in periodic pulses instead of continuously. Another approach is the expert-trainee paradigm in which the expert performs movements, which are followed by the trainee via haptic feedback based on the kinematic errors between the expert and trainee .
Haptic wearables have the potential to address sensory impairments. We define haptics broadly as the sense of touch and includes vibration, texture, slip, temperature, pain, force and proprioception sensations. Smaller, lighter, and more powerful sensors, actuators, and processors have enabled a recent rise in wearable technology for clinical applications. Wearable systems have been used for performing home rehabilitation, assessing functional activity, detecting movement disorders, improving walking stability, and reducing joint loading [15–17]. These systems give users mobility and the freedom to perform normal tasks in natural environments.
Due to recent rapidly increasing interest in wearables for clinical, research, and commercial purposes, there is a need to clearly present the state-of-the-art as it relates to impairments and rehabilitation. Thus, the purpose of this review is to examine haptic wearables for applications of varying degree of sensory impairment. While the focus was on portable devices, tethered devices demonstrating clinical benefits of wearable haptic feedback that could be made portable (e.g. battery-powered instead of outlet-powered) were also included. Wearable robotic rehabilitation or powered exoskeleton devices were not included as they have been the subject of previous review [21, 22]. The paper is organized by descending degree of sensory impairment beginning with sensory replacement, then sensory augmentation, and finally trainer.
Haptic wearables can act as a sensory replacement for total impairments. This section covers haptic applications involving missing upper and lower limbs followed by vision and auditory loss.
Prosthetic hands have achieved remarkable mechatronic capabilities (e.g. Revolutionizing Prosthetics and Otto Bock), however, up to 39 % of amputees wearing myoelectrically controlled prostheses do not use them regularly or at all due to a lack of tactile sensory feedback [23–26]. Current grasp information in prosthetic users occurs through visual observation (77 %), listening (67 %) and residual limb sensations (57 %) . Haptics for total impairment aims to restore missing tactile or proprioceptive information vital to prosthetic grasp to prolong sustained prosthesis use [28–31]. A major challenge is orchestrating spatial and temporal stimulation patterns and energy demands such that they give rise to congruent neuronal representations of vibration, contact, force, pressure, slip or muscle impedance during long-term use.
Slip, or shear forces between prosthesis and object held, is pivotal for determining grasp stability and minimum grasp force [40–44]. Slip and force feedback in combination allow manipulation of a virtual object with lower forces than with force feedback alone . Slip speed feedback, implemented as electrotactile stimulation on the skin, increases the success in stopping slip and regulates the user’s grip reaction time . Kim et al.  built a tactile device for amputees after targeted nerve reinnervation surgery (Fig. 2 (right)). The device relays contact, pressure, vibration and shear through a mechanically-actuated tactor in contact with an 8 mm diameter patch of skin. Damian et al.  developed a wearable haptic device that relays slip speed, through a series of tactors that sweep across the skin and grip force through frequency-encoded tapping on the skin.
While many skin sites have been explored for tactile stimulation [49–52], fingertips are an attractive location due to the high density of the mechanoreceptors and the congruency of grasp sensation with the lost hand. Sites closest to the lost limb are preferred for the exploitation of redundant afferent terminals [35, 48, 53]. Other locations where skin sensation is used relatively less in normal life such as the arm or back have a lower density of mechanoreceptors but do not interfere with manipulative tasks [33, 52, 54]. However, it may be that the location of skin stimulation is less important than other factors such as learning rates .
Artificial motion proprioception allows prosthesis users to reach targets more accurately and reduces visual attention during manipulation [56, 57]. Witteveen et al.  used an array of eight vibrotactors on the arm to represent eight discrete positions in closing a prosthetic hand during grasping. Vibrotactile feedback was found superior to no feedback in grasp success and duration during virtual object grasping tasks. Bark et al.  introduce a wearable haptic device for rotational skin stretch to display proprioceptive limb motion. Users were able to discriminate rotational displacements of stretch within 6 degrees of the total range of motion. Artificial impedance feedback can support prostheses users to adapt the interaction of their prosthesis to a variety of environments. Blank et al.  showed that human users provided with position and force feedback are able to evaluate the effects of prosthesis impedance and its adjustability improves the users’ performance in minimizing contact forces with a moving object. In addition, vibrotactile  and skin stretch  have been used to provide users with the ability to regulate environment interaction forces.
These investigations show clear benefits of wearable haptic feedback for upper-limb prosthetics by restoring lost force, pressure, slip, and proprioception sensations. Current studies have primarily focused on restoring a single sensation, such as slip, while restoring multiple sensations simultaneously could endow users with more stable grasp and higher dexterity in real-life manipulation scenarios. A major challenge is miniaturizing bulky multi-function haptic wearables to a size where the benefits of the wearable device outweigh discomfort and inconveniences of complex devices which have thus far limited long-term user compliance.
While a variety of lower limb prostheses exist, relatively few provide sensory feedback as compared to upper limb prosthetics . However, the absence of feedback can lead to abnormalities in gait coordination, deficient balance, and prolonged rehabilitation [63–65]. To relay ground-to-prosthesis contact force information, Fan et al.  developed a tactile system consisting of a cuff of four silicone pneumatic balloons placed around the thigh that respond monotonically to pressure patterns recorded by force sensors in the insole of the user. Six healthy subjects were able to differentiate inflation patterns and direction of pressure stimuli, recognize three force levels and discriminate gait movements with 99.0 %, 94.8 %, 94.4 % and 95.8 % accuracy, respectively. Crea et al.  mapped the force recorded in the insole to vibrotactile feedback on the thigh skin, providing information about gate-phase transition. They demonstrated that the spatial and temporal relationships between vibrotactile time-discrete feedback and gait-phase transitions can be learned. In a study on twenty four transtibial prostheses users, Rusaw et al.  conveyed body motion through vibratory feedback proportional to signals from force sensors placed under the prosthetic foot. Vibratory feedback improved postural stability and reduced response time for avoiding falls. Proprioceptive feedback in lower-limb prostheses was investigated by Buma et al.  using a spatial electrotactile display of the prosthetic knee angle during gait. Subjects wore electrodes on the medial side of the thigh just above the knee, and the results showed that intermittent stimulation reduced habituation after 15 minutes. Finally, Sharma et al.  investigated the response in limb motion given vibration stimuli applied to the thigh, and showed that average response time was 0.8 sec, and response accuracy was greater than 90 %.
Most studies involving wearable haptics for lower-limb prosthetics have extracted various gait characteristics, such as foot pressure patterns or gait phase detection, from force-sensing insoles and then mapped these characteristics to prosthetic users via haptic feedback. While these initial studies are promising, future research should focus on restoring missing proprioceptive sensations at the ankle and knee joints in combination with foot pressure patterns.
Vision aid for the blind
Engineers and scientists have long sought to enable visual substitution for the blind. In a seminal study, Bach-Y-Rita et al.  used a 20 x 20 array of tactors embedded in a dental chair to stimulate the skin of the back of blind subjects giving them a sense of “vision” through tactile substitution. Research built on these initial efforts has resulted in a host of haptic wearables as vision aids for the blind (see survey articles [72, 73]).
Although the waist has low tactile acuity, it is a natural location for haptic feedback as it moves relatively little during ambulation. McDaniel et al.  developed a tactile belt of 7 equidistantly spaced tactors around the waist to cue a blind user of another person’s presence. Results showed that the belt could convey another person’s direction via vibration location and another person’s distance via vibration duration. Karcher et al.  used a tactile belt consisting of 30 equidistantly spaced tactors in combination with a digital compass to display the direction of magnetic north by continually vibrating the closest tactor aligned with the magnetic north direction. Johnson and Higgins  used a tactile belt with two attached web cameras to convert visual information to a two-dimensional tactile depth map. Sensed objects triggered belt vibrations in the object’s direction, with closer objects causing higher vibration frequencies. Several studies have used tactile belts with GPS sensing for outdoor navigation by vibrating tactors in the direction of required movement to reach an intended waypoint or final destination [77–79].
There is a clear tradeoff between user comfort and density of feedback information when deciding on the location to apply haptic feedback as a vision aid. While applying tactile sensations to the waist or sole of the foot may be natural locations given that most people already wear belts and shoe insoles, stimulating high-density mechanoreceptor areas such as the mouth and fingertips enables higher resolution feedback that may more realistically convey visual information. A key emphasis moving forward should be identifying the most critical visual information for the blind and mapping this in an intuitive way to the users. Given that human response to visual information tends to be application specific, such as responding to non-verbal communication cues versus changing gait patterns to avoid an identified obstacle during navigation, haptic feedback strategies may also need to be application-specific instead of attempting to generalize all visual information.
Auditory aid for the deaf
To hold conversations, the hearing impaired typically rely on visual or tactile cues, such as fingerspelling, lip reading, or Tadoma. Alternatively, tactile vocoders perform a frequency analysis of incoming auditory signals and display spectral information as stimulation on the skin of the hearing impaired [91, 92]. Saunders et al.  presented an abdomen belt of electrotactile stimulators encoding speech frequencies for speech recognition in profoundly deaf children (hearing loss of greater than 90 dB for 250 Hz sound frequencies). Improvement in speech production and intelligibility was observed after a 4-month exploratory study. Boothroyd et al.  showed that intonation can be more easily recognized using mechanical strokes on the skin implemented as an array of eight solenoids actuated depending on the pitch extracted from a microphone or accelerometer. A comparison between multichannel vibrotactile and electrical tactile stimulation for relaying sound frequency is presented in . The two tactile display devices differed in stimulation modality (vibrotactile, electrotactile), location of stimulation (forearm, abdomen), and voice processing (with and without noise suppression). Results showed that both devices provide benefits beyond lipreading alone. Bernstein et al.  compared three vibrotactile vocoders on the forearm in normal and hearing-impaired subjects and found that greater resolution in the second formant region and linear output scaling led to significant improvements of sentence lipreading with vocoders.
Apart from speech recognition, it is also difficult for the hearing impaired to discriminate environmental sound. Reed et al.  demonstrated that normal hearing and profoundly deaf subjects equipped with a wearable spectral tactual aid are able to identify two bits of information in four 10-item sets of sounds. Furthermore, because it is difficult for the hearing-impaired to control voice pitch, it is challenging for them to maintain a stable tone while speaking or singing. Sakajiri et al.  developed a device of 64 piezoelectric vibrators arranged in rows of displacing pins that contact the user’s finger. The pins push onto the skin displaying the difference between user and target pitch. Two hearing-impaired subjects with knowledge and practice in music tested the device capability to aid their singing. The tactile display system reduces the average musical interval deviation to 117.5 cent (cent is a logarithmic unit of measure used for musical intervals), which is comparable to that of normal hearing children.
The inherent complexity of language and subject-to-subject differences raises serious challenges in developing highly effective haptic displays for auditory replacements. It may be more realistic for haptic feedback to supplement existing auditory activities such as supplementing lipreading to resolve ambiguous lip-read messages [96, 99]. Further research should integrate more sensed auditory modalities into wearable haptic technology, such as audio frequencies, voice aspiration, and temporal characteristics patterns. Further work to optimize voice signal filters to comply with subject-specific impairments could bring further benefits through haptic displays.
For partial sensory impairments, wearable haptics may provide complementary information to augment weak and noisy sensory signals. This section covers wearable haptics for improving standing balance, walking balance, and rehabilitation for varied conditions such as vestibular loss, Parkinson’s disease, and stroke.
Vibrotactile sensations are typically used as a repulsive instructional cue (i.e. move away from the vibration) , though attractive instructional cues might be compatible with non-volitional responses to vibrotactile stimulation over certain anatomical regions [106, 107]. Haggerty et al.  tested the effect of the attentional load of vibration feedback by requiring subjects to perform a secondary task during standing posture vibration training. Ten healthy older adults performed standing balance training while simultaneously performing a secondary cognitive task (identifying a high or low pitched tone either verbally or by pressing one of two buttons). Subjects improved postural stability while performing a secondary task though their response times increased suggesting that vibrotactile feedback can be used to improve postural stability for older adults in cognitive loading situations. While tactile feedback is typically given based on trunk kinematic measurements, it has recently been suggested that incorporating muscle activation measurements in combination with kinematics may be more effective .
While haptic feedback for posture sway training is usually applied to the torso, the head and tongue are also suitable stimulation locations [110, 111]. Vuillerme et al.  used a 6 x 6 array (overall size of 1.5 cm × 1.5 cm) of electrotactile electrodes (1.4 mm diameter) to map foot center-of-pressure measurements to the tongue. The location of electrode stimulation corresponds to the location of the center of foot pressure thus augmenting each subject’s foot center-of-pressure perception. Tongue tactile feedback has been used for standing posture rehabilitation in individuals with unilateral and bilateral areflexia and unilateral and bilateral vestibular losses .
In contrast with previous studies utilizing haptic wearables as a cueing-based response for altering users of desired movement changes, stochastic resonance tactile vibrations have been suggested to amplify natural human afferent signals by adding white noise to a weak signal [114, 115]. Priplata et al.  used gel-based insoles with three embedded tactors to apply stochastic resonance white noise vibration to the sole of the foot. Twenty-seven elderly subjects stood quietly on insoles in conditions with and without input white noise. The amplitude of the noise was set to 90 % of the sensory perception threshold for each subject (and thus the noise signal was imperceptible during testing), and noise frequencies were 0–100 Hz. All standing balance metrics improved with stochastic noise. A similar study was performed showing stochastic resonance also improves standing balance for individuals with diabetic neuropathy and stroke .
Two primary strategies have emerged for applying wearable haptic feedback to augment standing balance: 1) apply periodic tactile cues, often to the torso, to instruct a desired corrective movement, and 2) apply continuous vibrations to the foot sole to amplify natural afferent signals. Combining these two methods could enable a superior system with greater potential to improve balance. Additionally, most studies assume wearable haptic devices need to be used indefinitely to continue providing balance aid benefits, while ignoring the effects of long-term learning and adaptation to such devices, which is a critical aspect deserving future consideration.
Trunk movement in the medial-lateral plane is crucial for postural stability during gait . Thus, research efforts have focused on providing tactile feedback to reduce excessive medial-lateral trunk movements. Dozza et al.  used a vibrotactile vest for gait training in nine subjects with unilateral vestibular loss. The vest contained two columns of three tactors on each side and pairs vibrated when medial-lateral trunk tilt exceeded 2 degrees (lower pair), 7 degrees (middle pair), and 12 degrees (higher pair). This training resulted in reduced trunk tilt, center of mass displacement, medial-lateral step width, and frequency of stepping error during gait. Horak et al.  performed two tactile feedback training sessions spaced two weeks apart in 10 individuals with unilateral vestibular loss. Feedback increased walking stability during tandem gait (heel-to-toe walking) evidenced by reductions in center-of-mass displacement, trunk tilt, and medial–lateral step width. Janssen et al.  tested 40 healthy subjects and showed that a vibrotactor visor utilizing tactile, visual, and auditory feedback reduced trunk tilt velocity and angles for a variety of gait tasks including walking: with eyes open or closed, while rotating or pitching the head, while carrying a glass of water, backwards, and up and down stairs.
Tactile feedback can increase attentional load during gait. Verhoeff et al.  observed 16 healthy young and 13 healthy old subjects as they performed gait training with a simultaneous secondary task, either walking while counting backwards in 7’s (cognitive task) or walking while carrying a tray with cups of water (motor task). Young subjects were able to perform both dual tasks, but elderly subjects could only perform the dual motor task and not the dual cognitive task. In gait retraining, continuous vibration feedback may be more appropriate than short periodic vibration pulses. Sienko et al.  tested seven subjects with vestibular loss who received either continuous vibration feedback of their trunk tilt angle or a periodic 200 ms vibration pulse immediately following heel strike on each step. While both methods enabled subjects to reduce medial-lateral trunk sway, continuous feedback was more effective.
Similar to applications in standing balance, stochastic resonance has been proposed as sensory augmentation to boost weak afferent signals for gait. Galica et al.  inserted three tactors into customized sandals to deliver 0–100 Hz white noise to 18 elderly recurrent fallers and 18 elderly non-fallers during 1 m/s walking gait. White noise foot vibrations reduced stride, stance, and swing time variability for elderly recurrent fallers and reduced stride and stance time variability for elderly non-fallers.
The benefits of wearable haptic feedback during gait must be weighed against the potential drawbacks. While tactile cues can help improve balance by reducing trunk sway, they also require additional cognitive attention that could result in negative secondary effects such as missing a curb while walking across a street. Future work should implement wearable haptic training systems that seek to minimize attentional load while maximizing gait improvements.
For patients with neurological diseases, such as stroke, Parkinson’s disease, spinal cord injury, and peripheral neuropathy, haptic sensation is lost or distorted making everyday tasks difficult . Artificial haptic feedback can play a role in regaining lost motor control . Motor function improvement is achieved through task-oriented repetitive training during functionally related dynamic movements and the provision of artificial feedback [125, 126].
Upper extremity rehabilitation is often performed via vibrotactile feedback applied to the arm or hand to guide limb movements [8, 9, 36, 100, 127]. Jiang et al.  built a tactile wearable device to help multiple sclerosis patients improve grasp force during manipulation tasks by transmitting tactile information as a vibrotactile signal on the fingernail. Amplitude-based vibrotactile feedback was useful for patients with mild impairment in alerting them when grip force exceeded a predefined threshold. For those with severe impairment, better results were achieved by providing a feedback signal in which the frequency and duty cycle were proportional to the magnitudes of the contact forces. Lieberman et al.  developed a 5-DOF wearable robotic suit for improving human motion learning in rehabilitation. The suit was equipped with vibrotactile actuators placed near body joints which encoded arm postures. Tactile feedback provided by the suit yielded a 27 % improvement in accuracy while performing the target motion, and an accelerated learning rate of up to 23 %, compared to no feedback.
While rehabilitation studies show increased performance with tactile feedback, a major disadvantage remains the variability between subjects, which impedes finding optimal feedback standards. Rehabilitation platforms capable of intelligent, adaptable tactile feedback configurations could provide subject-specific treatment more universally useful.
While most haptic trainer studies have not been clinically focused (e.g. drumming  or snowboarding  and jump landings ), increasing interest in haptic wearables makes this a likely area of growth. For example, haptic wearables can reduce knee loads by providing motion cues that alter risky walking patterns. One approach is to give subjects haptic feedback information directly related to knee loading and allow them to self-select a new gait pattern to reduce knee loads. Wheeler et al.  attached a single vibrotactor to the forearm which vibrated when knee loads exceeded a predefined threshold. No feedback was given when new gait patterns resulted in lower knee loads. Although effective in short-term, one drawback of this method is that subjects often self-selected awkward gait patterns that would likely not be maintained long-term.
Another approach is to explicitly train gait kinematics to reduce knee loading. Dowling et al.  embedded a pager motor inside a shoe to give vibration feedback to the foot based on lateral foot pressure. On each step subjects walked with lateral foot pressure above a predefined threshold, measured with a force-sensing resistor on the lateral underside of the shoe, the pager motor vibrated instructing a change in gait. Subjects quickly learned the medial foot pressure gait patterns, which resulted in significantly reduced knee loads. In other studies, vibration pulses on the lateral aspect of the shank just below the knee have been used to train individuals with knee osteoarthritis to internally rotate their toes by 5–7 degrees resulting in reduced knee loading and reduced knee pain over time [138, 139].
Training multiple kinematic parameters simultaneously  presents cognitive and motor challenges related to receiving and responding to multiple simultaneous channels of information. Lurie et al.  trained subjects to walk with new gait patterns involving kinematic changes to trunk sway, tibia angle, and foot progression angle by either giving error correction feedback cues on all parameters simultaneously or one parameter at a time. Perception accuracy was lower when all three vibrations were presented simultaneously on three consecutive steps as compared to one distinct vibration on each of the three steps. Subject performance was the same for all tactile feedback simultaneously and one feedback parameter at a time despite the fact that less feedback information was transmitted in the one feedback per step scenario. In another study, Jirattigalachote et al.  showed that when presenting multiple tactile feedback channels at separate skin locations simultaneously, subjects more accurately perceive different haptic stimuli (e.g. fast-adapting mechanoreceptor activation at one location and slow-adapting mechanoreceptor activation at the other location) compared to alike haptic stimuli during standing, walking, and jogging.
While haptic wearables have generally focused on treating existing problems, a shift in focus towards preventative medicine could enable a greater depth and impact in clinical applications. Knee osteoarthritis is one application in which tactile feedback has already been used to retrain gait movements to reduce knee loads that could potentially prevent the future development of osteoarthritis. Other future applications of wearable haptics as trainer could include correcting sitting posture to prevent back and neck injuries or correcting athletic movements to prevent ligament tears or bone fractures.
For patients with total sensory impairment, haptic wearables can transmit missing information related to manipulation, walking, or speaking to complete the otherwise broken sensorimotor control loop. Motor disorders associated with partial sensory impairment have been addressed with haptic wearables that transmit behavioral cues, such as posture and gait guidance based on kinematic error signals in specific rehabilitation tasks. This same approach can be used for people with no sensory impairment to instruct movement changes to improve performance or prevent injury or disease. In addition to the specific suggestions for future work presented in each previous individual section in the body of this paper, we identified the following general design principles, based on the reviewed studies, important for developing future wearable haptic systems for sensory impairment:
From need to practice. A practical and efficient development of haptic wearables should follow a rigorous identification of the clinical requirements of the target condition. Haptic wearables must be collaboratively and comprehensively developed by involving clinicians, patients, scientists, and engineers, such that the devices are a product of clinical observations, direct end-user evaluation and feedback, up-to-date and integrative scientific knowledge and wearable technology [24, 25, 28].
Bioelectrical/biomechanical compliance. While various systems have been explored that demonstrate successful haptic mapping, further work is needed to develop mechanisms for long term efficacy and wearability, with special attention taken to comply with user kinematics, avoid user pain and fatigue, . Reduced prosthesis weight has been found to be the highest priority design concern of prostheses users . Miniature soft actuators [143–145] could ensure light haptic devices that do not impede the natural motions of the human body where they are mounted.
Intuitive multimodal haptic representation. The haptic representation of the transmitted information must be intuitive and easy to use . Depending on the sensory impairment, haptic signals can display mechanics information (e.g. forces or angles) or instructional cues (e.g. desired movement change) encoded by signal magnitude, frequency or location on the skin. This pursuit becomes more challenging as multimodal feedback is integrated. Although most studies have only focused on a single modality, integrating multiple haptic modalities is necessary to comprehensively compensate for the missing sensation, e.g., force and slip feedback for upper extremity prosthetic manipulation, and limb position and planar pressure feedback for walking rehabilitation.
Low energy demands. Long term wearables rely on sustainable actuation and sensing. Novel energy sources and energy management should be considered in the design of the haptic device [147, 148]. For example, careful selection of power sources with high power-to-weight ratios and on-board computational algorithms to minimize power consumption could help meet these demands for tasks requiring extensive user training and long-term use.
Long term usage. Most haptic wearables are currently tested in short term tasks under laboratory conditions. Long-term testing is critical for developing and assessing sustainable haptic devices. This pursuit could significantly affect wearable device design and the implementation of feedback schemes and adaptive control algorithms to maintain the user performance over time.
One persistent question that repeatedly arose was, are haptic wearables best suited as temporary or permanent devices? Temporary devices can be used to train new movements which would eventually be internalized. Conversely, permanent feedback devices would be used indefinitely much like a prosthesis . Horak et al.  showed that gait stability learning from biofeedback was not retained when the biofeedback was removed for a tandem gait task, and Dozza et al.  showed that a single session of practice with feedback did not result in lasting after-effects, which both indicate the need for either long-term training or permanent use. The duration of haptic wearables use may depend on the severity of the sensory impairment and the ability for long-term, sustainable motor learning in target populations. Ultimately, the fundamental goal of the haptic wearables is to assist sensory impairments in an unobtrusive manner, regardless of the severity of the user’s condition or length of treatment [149, 150].
This work was supported by the University of Michigan–Shanghai Jiao Tong University Collaboration on Nanotechnology for Energy and Biomedical Applications (14X120010006), the Swiss National Science Foundation (PBZHP2 143344), the National Basic Research Program (973 Program) of China (2011CB013305), and the National Natural Science Foundation of China (51121063).
- Kusoffsky A, Wadell I, Nilsson L. The relationship between sensory impairment and motor recovery in patients with hemiplegia. Scand J Rehabil Med. 1982;14:27–32.PubMedGoogle Scholar
- Stern PH, Mcdowell F, Miller JM, Robinson M. Factors influencing stroke rehabilitation. Stroke. 1971;2:213–8.PubMedGoogle Scholar
- Lindenberger U, Ghisletta P: Cognitive and sensory declines in old age: Gauging the evidence for a common cause. Psychol Aging. 2009; 24:1-16.Google Scholar
- Geldard F. Adventures in tactile literacy. Am Psychol. 1957;12:115–24.Google Scholar
- Jones LA, Sarter NB. Tactile Displays: Guidance for Their Design and Application. Hum Factors J Hum Factors Ergon Soc. 2008;50:90–111.Google Scholar
- Bark K, Wheeler J, Shull PB, Savall J, Cutkosky M. Rotational skin stretch feedback: a wearable haptic display for motion. IEEE Trans Haptics. 2010;3:166–76.Google Scholar
- Antfolk C, D’Alonzo M, Rosén B, Lundborg G, Sebelius F, Cipriani C. Sensory feedback in upper limb prosthetics. Expert Rev Med Devices. 2013;10:45–54.PubMedGoogle Scholar
- Lieberman J, Breazeal C. TIKL: development of a wearable vibrotactile feedback suit for improved human motor learning. IEEE Trans Robot. 2007;23:919–26.Google Scholar
- Kapur P, Jensen M, Buxbaum LJ, Jax SA., Kuchenbecker KJ: Spatially distributed tactile feedback for kinesthetic motion guidance. In 2010 IEEE Haptics Symp. Proc. IEEE; 2010;519–526.Google Scholar
- Wall C, Kentala E. Effect of displacement, velocity, and combined vibrotactile tilt feedback on postural control of vestibulopathic subjects. J Vestib Res Equilib Orientat. 2010;20:61–9.Google Scholar
- Bächlin M, Förster K, Tröster G. SwimMaster: A Wearable Assistant for Swimmer. In: Proc 11th Int Conf Ubiquitous Comput - Ubicomp ’09. New York, New York, USA: ACM Press; 2009. p. 21.Google Scholar
- Förster K, Bächlin M, Tröster G. Non-interrupting user Interfaces for Electronic Body-worn Swim Devices. In: Proc 2nd Int Conf PErvsive Technol Relat to Assist Environ - PETRA ’09. New York, New York, USA: ACM Press; 2009. p. 1–4.Google Scholar
- Shull PB, Lurie K, Cutkosky MR, Besier T. Training multi-parameter gaits to reduce the knee adduction moment with data-driven models and haptic feedback. J Biomech. 2011;44:1605–9.PubMedGoogle Scholar
- Lee B-C, Chen S, Sienko KH. A wearable device for real-time motion error detection and vibrotactile instructional cuing. IEEE Trans Neural Syst Rehabil Eng. 2011;19:374–81.PubMedGoogle Scholar
- Patel S, Park H, Bonato P, Chan L, Rodgers M. A review of wearable sensors and systems with application in rehabilitation. J Neuroeng Rehabil. 2012;9:21.PubMed CentralPubMedGoogle Scholar
- Steins D, Dawes H, Esser P, Collett J. Wearable accelerometry-based technology capable of assessing functional activities in neurological populations in community settings: a systematic review. J Neuroeng Rehabil. 2014;11:1–13.Google Scholar
- Shull PB, Jirattigalachote W, Hunt MA, Cutkosky MR, Delp SL. Quantified self and human movement: a review on the clinical impact of wearable sensing and feedback for gait analysis and intervention. Gait Posture. 2014;40:11–9.PubMedGoogle Scholar
- Bach-y-Rita PW, Kercel S. Sensory substitution and the human–machine interface. Trends Cogn Sci. 2003;7:541–6.PubMedGoogle Scholar
- Dozza M, Wall C, Peterka RJ, Chiari L, Horak FB. Effects of practicing tandem gait with and without vibrotactile biofeedback in subjects with unilateral vestibular loss. J Vestib Res. 2007;17:195–204.PubMed CentralPubMedGoogle Scholar
- Sigrist R, Rauter G, Riener R, Wolf P. Augmented visual, auditory, haptic, and multimodal feedback in motor learning: a review. Psychon Bull Rev. 2013;20:21–53.PubMedGoogle Scholar
- Reinkensmeyer DJ, Boninger ML. Technologies and combination therapies for enhancing movement training for people with a disability. J Neuroeng Rehabil. 2012;9:17.PubMed CentralPubMedGoogle Scholar
- Tefertiller C, Pharo B, Evans N, Winchester P. Efficacy of rehabilitation robotics for walking training in neurological disorders: A review. J Rehabil Res Dev. 2011;48:387.PubMedGoogle Scholar
- Biddiss EA, Chau TT. Upper limb prosthesis use and abandonment: a survey of the last 25 years. Prosthet Orthot Int. 2007;31:236–57.PubMedGoogle Scholar
- Pylatiuk C, Schulz S, Döderlein L. Results of an Internet survey of myoelectric prosthetic hand users. Prosthet Orthot Int. 2007;31:362–70.PubMedGoogle Scholar
- Biddiss E, Beaton D, Chau T. Consumer design priorities for upper limb prosthetics. Disabil Rehabil Assist Technol. 2007;2:346–57.PubMedGoogle Scholar
- Østlie K, Lesjø IM, Franklin RJ, Garfelt B, Skjeldal OH, Magnus P. Prosthesis rejection in acquired major upper-limb amputees: a population-based survey. Disabil Rehabil Assist Technol. 2014;7:1–2.Google Scholar
- Lewis S, Russold MF, Dietl H: User demands for sensory feedback in upper extremity prostheses. In 2012 IEEE International Symp. on Medical Meas. and Appl. Proceedings (MeMeA). IEEE; 2012;1–4.Google Scholar
- Peerdeman B, Boere D, Witteveen H, Huis in 'tVeld R, Hermens H, Stramigioli S, et al. Myoelectric forearm prostheses: State of the art from a user-centered perspective. J Rehabil Res Dev. 2011;48:719.PubMedGoogle Scholar
- Atkins DJ, Heard DCY, Donovan WH. Epidemiologic overview of individuals with upper-limb loss and their reported research priorities. J Prosthetics Orthot. 1996;8:1–11.Google Scholar
- Saunders I, Vijayakumar S. The role of feed-forward and feedback processes for closed-loop prosthesis control. J Neuroeng Rehabil. 2011;8:60.PubMed CentralPubMedGoogle Scholar
- Hernandez Arieta A, Dermitzakis K, Damian D, Lungarella M, Pfeifer R: Sensorymotor coupling in rehabilitation robotics. In Serv Robot Appl. Edited by Yoshihiko Takahashi (Ed.). INTECH Open Access Publisher, 2008;21–36.Google Scholar
- Meek SG, Jacobsen SC, Goulding PP. Extended physiologic taction: design and evaluation of a proportional force feedback system. J Rehabil Res Dev. 1989;26:53–62.PubMedGoogle Scholar
- Patterson PE, Katz JA. Design and evaluation of a sensory feedback system that provides grasping pressure in a myoelectric hand. J Rehabil Res Dev. 1992;29:1.PubMedGoogle Scholar
- Antfolk C, Balkenius C, Lundborg G, Rosén B, Sebelius F. Design and technical construction of a tactile display for sensory feedback in a hand prosthesis system. Biomed Eng Online. 2010;9:50.PubMed CentralPubMedGoogle Scholar
- Antfolk C, Alonzo MD, Controzzi M, Lundborg G, Rosén B, Sebelius F, et al. Artificial redirection of sensation from prosthetic fingers to the phantom hand map on transradial amputees: Vibrotactile versus mechanotactile sensory feedback. IEEE Trans Neural Syst Rehabil Eng. 2013;21:112–20.PubMedGoogle Scholar
- Jiang L, Cutkosky M, Ruutiainen J, Raisamo R. Using haptic feedback to improve grasp force control in multiple sclerosis patients. IEEE Trans Robot. 2009;25:593–601.Google Scholar
- Brown JD, Paek A, Syed M, O’Malley MK, Shewokis PA, Contreras-Vidal JL, Davis AJ, Gillespie RB: Understanding the role of haptic feedback in a teleoperated/prosthetic grasp and lift task. World Haptics Conf 2013:271–276.Google Scholar
- Tejeiro C, Stepp CE, Malhotra M, Rombokas E, Matsuoka Y: Comparison of remote pressure and vibrotactile feedback for prosthetic hand control. Proc IEEE RAS EMBS Int Conf Biomed Robot Biomechatronics 2012;521–525Google Scholar
- Rombokas E, Stepp CE, Chang C, Malhotra M, Matsuoka Y. Vibrotactile sensory substitution for electromyographic control of object manipulation. IEEE Trans Biomed Eng. 2013;60:2226–32.PubMedGoogle Scholar
- Augurelle A-S, Smith AM, Lejeune T, Thonnard J-L. Importance of cutaneous feedback in maintaining a secure grip during manipulation of hand-held objects. J Neurophysiol. 2003;89:665–71.PubMedGoogle Scholar
- Johansson RS, Westling G. Roles of glabrous skin receptors and sensorimotor memory in automatic control of precision grip when lifting rougher or more slippery objects. Exp Brain Reseash. 1984;56:550–64.Google Scholar
- Johansson RS, Cole KJ. Grasp stability during manipulative actions. Can J Physiol Pharmacol. 1994;72:511–24.PubMedGoogle Scholar
- Tsagarakis NG, Horne T, Caldwell DG: Slip aestheasis: A portable 2D slip/skin stretch display for the fingertip. First Jt Eurohaptics Conf Symp Haptic Interfaces Virtual Environ Teleoperator Syst 2005:214–219Google Scholar
- Walker JM, Blank AA, Shewokis PA, O’Malley MK. Tactile feedback of object slip improves performance in a grasp and hold task. IEEE Haptics Symp. 2014;2014:461–6.Google Scholar
- Webster RJ, Murphy TE, Verner LN, Okamura AM. A novel two-dimensional tactile slip display: design, kinematics and perceptual experiments. ACM Trans Appl Percept. 2005;2:150–65.Google Scholar
- Damian DD, Arita AH, Martinez H, Pfeifer R. Slip speed feedback for grip force control. IEEE Trans Biomed Eng. 2012;59:2200–10.PubMedGoogle Scholar
- Kim K, Member A, Colgate JE, Santos-munn JJ, Makhlin A, Peshkin MA. On the design of miniature haptic devices for upper extremity prosthetics. IEEE/ASME Trans on Mechatronics. 2010;15:27–39.Google Scholar
- Damian DD, Ludersdorfer M, Kim Y, Hernandez Arieta A, Pfeifer R, Okamura AM. Wearable haptic device for cutaneous force and slip speed display. IEEE Int Conf Robot Autom. 2012;2012:1038–43.Google Scholar
- Bach-y-Rita P, Kaczmarek KA, Tyler ME, Garcia-Lara J. Form perception with a 49-point electrotactile stimulus array on the tongue: a technical note. J Rehabil Res Dev. 1998;35:427–30.PubMedGoogle Scholar
- Panarese A, Edin BB, Vecchi F, Carrozza MC, Johansson RS. Humans can integrate force feedback to toes in their sensorimotor control of a robotic hand. IEEE Trans Neural Syst Rehabil Eng. 2009;17:560–7.PubMedGoogle Scholar
- Wheeler J, Bark K, Savall J, Cutkosky M. Investigation of rotational skin stretch for proprioceptive feedback with application to myoelectric systems. IEEE Trans neural Syst Rehabil Eng. 2010;18:58–66.PubMedGoogle Scholar
- Seps M, Dermitzakis K, Hernandez-arieta A: Study on lower back electrotactile stimulation characteristics for prosthetic sensory feedback. In 2011 IEEE/RSJ Int Conf Intell Robot Syst; 2011:3454–3459.Google Scholar
- Brown JD, Gillespie RB, Gardner D, Gansallo E a. Co-location of force and action improves identification of force-displacement features. 2012 IEEE Haptics Symp 2012:187–193.Google Scholar
- Van Der Riet D, Stopforth R, Bright G, Diegel O: Simultaneous vibrotactile feedback for multisensory upper limb prosthetics. Proc - 2013 6th Robot Mechatronics Conf RobMech 2013 2013:64–69.Google Scholar
- Stepp CE, Matsuoka Y. Object manipulation improvements due to single session training outweigh the differences among stimulation sites during vibrotactile feedback. IEEE Trans Neural Syst Rehabil Eng. 2011;19:677–85.PubMed CentralPubMedGoogle Scholar
- Blank A, Okamura AM, Kuchenbecker KJ: Identifying the role of proprioception in upperlimb prosthesis control: Studies on targeted motion. ACM Transactions on Applied Perception. 2010;7(3), Article #15:1–19.Google Scholar
- Gurari N, Kuchenbecker KJ, Okamura AM. Perception of springs with visual and proprioceptive motion cues: Implications for prosthetics. IEEE Trans Human-Machine Syst. 2013;43:102–14.Google Scholar
- Witteveen HJB, Droog EA, Rietman JS, Veltink PH. Vibro- and electrotactile user feedback on hand opening for myoelectric forearm prostheses. IEEE Trans Biomed Eng. 2012;59:2219–26.PubMedGoogle Scholar
- Blank A, Okamura M, Whitcomb LL. Task-dependent impedance and implications for upper-limb prosthesis control. Int J Rob Res. 2014;33:827–46.Google Scholar
- Blank A, Okamura AM, Whitcomb LL: User comprehension of task performance with varying impedance in a virtual prosthetic arm: A pilot study. 4th IEEE RAS EMBS Int Conf Biomed Robot Biomechatronics 2012:500–507.Google Scholar
- Schorr SB, Quek ZF, Romano RY, Nisky I, Provancher WR, Okamura AM: Sensory substitution via cutaneous skin stretch feedback. In 2013 IEEE Int Conf Robot Autom; 2013:2341–2346.Google Scholar
- Martin J, Pollock A, Hettinger J. Microprocessor lower limb prosthetics: Review of current state of the art. J Orthotists annd Prosthetists. 2010;22:183–93.Google Scholar
- Gailey R. Rehabilitation of a traumatic lower limb amputee. Physiother Res Int. 2006;3:4–7.Google Scholar
- Lamoth CJC, Ainsworth E, Polomski W, Houdijk H. Variability and stability analysis of walking of transfemoral amputees. Med Eng Phys. 2010;32:1009–14.PubMedGoogle Scholar
- Wentink E, Talsma-Kerkdijk E, Rietman H, Veltink P: Feasibility of error-based electrotactile and auditive feedback in prosthetic walking. Prosthet Orthot Int. 2015;39:255-259.Google Scholar
- Fan RE, Culjat MO, King C-H, Franco ML, Boryk R, Bisley JW, et al. A haptic feedback system for lower-limb prostheses. IEEE Trans neural Syst Rehabil Eng. 2008;16:270–7.PubMedGoogle Scholar
- Crea S, Cipriani C, Donati M, Carrozza M, Vitiello N: Providing time-discrete gait information by wearable feedback apparatus for lower-limb amputees: usability and functional validation. IEEE Trans Neural Syst Rehabil Eng. 2015;23:250-257.Google Scholar
- Rusaw D, Hagberg K, Nolan L, Ramstrand N. Can vibratory feedback be used to improve postural stability in persons with transtibial limb loss? J Rehabil Res Dev. 2012;49:1239–54.PubMedGoogle Scholar
- Buma DG, Buitenweg JR, Veltink PH. Intermittent stimulation delays adaptation to electrocutaneous sensory feedback. IEEE Trans Neural Rehabil Eng. 2007;15:435–41.Google Scholar
- Sharma A, Torres-moreno R, Zabjek K, Andrysek J. Toward an artificial sensory feedback system for prosthetic mobility rehabilitation: Examination of sensorimotor responses. J Rehabil Res Dev. 2014;51:907–18.PubMedGoogle Scholar
- Bach-y-Rita P, Collins C, Saunders F, White B, Scadden L. Vision substitution by tactile image projection. Nature. 1969;221:963–4.PubMedGoogle Scholar
- Velazquez R. Wearable assistive devices for the blind. Wearable Auton Biomed Devices Syst Smart Environ. 2010;75:331–49.Google Scholar
- Dakopoulos D, Bourbakis NG. Wearable obstacle avoidance electronic travel aids for blind: A survey. IEEE Trans Syst Man Cybern. 2010;40:25–35.Google Scholar
- Mcdaniel T, Krishna S, Balasubramanian V, Colbry D, Panchanathan S: Using a haptic belt to convey non-verbal communication cues during social interactions to individuals who are blind. In HAVE Haptic Audio Vis Environ their Appl; 2008:1–6.Google Scholar
- Kärcher SM, Fenzlaff S, Hartmann D, Nagel SK, König P. Sensory augmentation for the blind. Front Hum Neurosci. 2012;6:1–15.Google Scholar
- Johnson L, Higgins CM. A navigation aid for the blind using tactile-visual sensory substitution. IEEE Eng Med Biol Conf. 2006;1:6289–92.Google Scholar
- Tsukada K, Yasumura M: ActiveBelt: Belt-type wearable tactile display. UbiComp; 2004:384–399.Google Scholar
- Van Erp JBF, van Veen HAHC, Jansen C, Dobbins T. Waypoint navigation with a vibrotactile waist belt. ACM Trans Appl Percept. 2005;2:106–17.Google Scholar
- Elliott LR, van Erp JBF, Redden ES, Duistermaat M. Field-based validation of a tactile navigation device. IEEE Trans Haptics. 2010;3:1–10.Google Scholar
- Amemiya T, Yamashita J, Hirota K, Michitaka H: Virtual leading blocks for the deaf-blind: A real-time way-finder by verbal-nonverbal hybrid interface and high-density RFID tag space. IEEE Virtual Real; 2004:165–173Google Scholar
- Meers S, Ward K: A substitute vision system for providing 3D perception and GPS navigation via electro-tactile stimulation. Int Conf Sens Technol; 2005:551–556Google Scholar
- Koo IM, Jung K, Koo JC, Nam J, Lee YK. Development of soft-actuator-based wearable tactile display. IEEE Trans Robot. 2008;24:549–58.Google Scholar
- Shah C, Bouzit M, Youssef M, Vasquez L: Evaluation of RU-Netra - tactile feedback navigation system for the visually impaired. Int Work Virtual Rehabil; 2006:72–77Google Scholar
- Ito K, Okamoto M, Akita J, Ono T, Gyobu I, Takagi T: CyARM: an alternative aid device for blind persons.CHI; 2005:1483–1486.Google Scholar
- Gallo S, Chapuis D, Santos-Carreras L, Kim Y, Retornaz P, Bleuler H, Gassert R: Augmented white cane with multimodal haptic feedback. 3rd IEEE RAS EMBS Int Conf Biomed Robot Biomechatronics 2010:149–155.Google Scholar
- Tang H, Beebe DJ. An oral tactile interface for blind navigation. IEEE Trans Neural Syst Rehabil Eng. 2006;14:116–23.PubMedGoogle Scholar
- Jones LA, Lockyer B, Piateski E. Tactile display and vibrotactile pattern recognition on the torso. Adv Robot. 2006;20:1359–74.Google Scholar
- Mann S, Huang J, Janzen R, Lo R, Rampersad V, Chen A, et al. Blind navigation with a wearable range camera and vibrotactile helmet. In: ACM Int Conf Multimed. New York, New York, USA: ACM Press; 2011. p. 1325–8.Google Scholar
- Velazquez R, Bazan O. Preliminary evaluation of podotactile feedback in sighted and blind users. IEEE Eng Med Biol Soc Conf. 2010;2010:2103–6.Google Scholar
- Menelas BJ, Otis MJ: Design of a serious game for learning vibrotactile messages. IEEE Haptic Audio Vis Environ Games; 2012:124–129.Google Scholar
- Kaczmarek K. Sensory Augmentation and Substitution. In: Bronzino EJD, editor. Biomed Eng Handb. 2nd ed. Boca Raton: CRC Press LLC; 2000.Google Scholar
- Rosen SM, Fourcin AJ, Moore BC. Voice pitch as an aid to lipreading. Nature. 1981;291:150–2.PubMedGoogle Scholar
- Saunders FA, Hill WA, Franklin B. A wearable tactile sensory aid for profoundly deaf children. J Med Syst. 1981;5:265–70.PubMedGoogle Scholar
- Boothroyd A. A wearable tactile intonation display for the deaf. IEEE Trans Acoust. 1985;33:111–7.Google Scholar
- Weisenberger JM, Broadstone SM, Saunders FA. Evaluation of two multichannel tactile aids for the hearing impaired. J Acoust Soc Am. 1989;86:1764–75.PubMedGoogle Scholar
- Bernstein LE, Demorest ME, Coulter DC, O’Connell M. Lipreading sentences with vibrotactile vocoders: Performance of normal-hearing and hearing-impaired subjects. J Acoust Soc Am. 1991;90:2971–84.PubMedGoogle Scholar
- Reed CM, Delhorne LA. The reception of environmental sounds through wearable tactual Aids. Ear Hear. 2003;24:528–38.PubMedGoogle Scholar
- Sakajiri M, Miyoshi S, Nakamura K, Fukushima S, Ifukube T: Voice pitch control using tactile feedback / or the deafblind or the hearing impaired persons to assist their singing. IEEE Int Conf Syst Man, Cybern; 2010:1483–1487.Google Scholar
- Yuan H, Reed CM, Durlach NI. Tactual display of consonant voicing as a supplement to lipreading. J Acoust Soc Am. 2005;118:1003–15.PubMedGoogle Scholar
- Gopalai AA, Senanayake SMNA. A wearable real-time intelligent posture corrective system using vibrotactile feedback. IEEE Trans Mechatronics. 2011;16:827–34.Google Scholar
- Kentala E, Vivas J, Wall C. Reduction of postural sway by use of a vibrotactile balance prosthesis prototype in subjects with vestibular deficits. Ann Otol Rhinol Laryngol. 2003;112:404–9.PubMedGoogle Scholar
- Wall C, Weinberg MS, Schmidt PB, Krebs DE. Balance prosthesis based on micromechanical sensors using vibrotactile feedback of tilt. IEEE Trans Biomed Eng. 2001;48:1153–61.PubMedGoogle Scholar
- Wall C, Weinberg MS: Balance prostheses for postural control. IEEE Eng Med Biol Mag 2003;22:84–90.Google Scholar
- Sienko KH, Balkwill MD, Oddsson LIE, Wall C. Effects of multi-directional vibrotactile feedback on vestibular-deficient postural performance during continuous multi-directional support surface perturbations. J Vestib Res. 2008;18:273–85.PubMedGoogle Scholar
- Jeka J, Lackner JR. Fingertip contact influences human postural control. Exp Brain Res. 1994;79:495–502.Google Scholar
- Lee B-C, Martin BJ, Sienko KH. Directional postural responses induced by vibrotactile stimulations applied to the torso. Exp brain Res. 2012;222:471–82.PubMedGoogle Scholar
- Lee B-C, Martin BJ, Ho A, Sienko KH. Postural reorganization induced by torso cutaneous covibration. J Neurosci. 2013;33:7870–6.PubMedGoogle Scholar
- Haggerty S, Jiang L-T, Galecki A, Sienko KH. Effects of biofeedback on secondary-task response time and postural stability in older adults. Gait Posture. 2012;35:523–8.PubMed CentralPubMedGoogle Scholar
- Honegger F, Hillebrandt IMA, van den Elzen NGA, Tang K-S, Allum JHJ. The effect of prosthetic feedback on the strategies and synergies used by vestibular loss subjects to control stance. J Neuroeng Rehabil. 2013;10:115.PubMed CentralPubMedGoogle Scholar
- Davis JR, Carpenter MG, Tschanz R, Meyes S, Debrunner D, Burger J, et al. Trunk sway reductions in young and older adults using multi-modal biofeedback. Gait Posture. 2010;31:465–72.PubMedGoogle Scholar
- Nanhoe-Mahabier W, Allum JH, Pasman EP, Overeem S, Bloem BR. The effects of vibrotactile biofeedback training on trunk sway in Parkinson’s disease patients. Parkinsonism Relat Disord. 2012;18:1017–21.PubMedGoogle Scholar
- Vuillerme N, Chenu O, Demongeot J, Payan Y. Controlling posture using a plantar pressure-based, tongue-placed tactile biofeedback system. Exp brain Res. 2007;179:409–14.PubMedGoogle Scholar
- Ghulyan-Bedikian V, Paolino M, Paolino F. Short-term retention effect of rehabilitation using head position-based electrotactile feedback to the tongue: Influence of vestibular loss and old-age. Gait Posture. 2013;38:777–83.PubMedGoogle Scholar
- Mcdonnell MD, Ward LM. The benefits of noise in neural systems: bridging theory and experiment. Nat Rev Neurosci. 2011;12:415–25.PubMedGoogle Scholar
- Moss F. Stochastic resonance and sensory information processing: a tutorial and review of application. Clin Neurophysiol. 2004;115:267–81.PubMedGoogle Scholar
- Priplata AA, Niemi JB, Harry JD, Lipsitz LA, Collins JJ. Vibrating insoles and balance control in elderly people. Lancet. 2003;362:1123–4.PubMedGoogle Scholar
- Priplata AA, Patritti BL, Niemi JB, Hughes R, Gravelle DC, Lipsitz LA, et al. Noise-enhanced balance control in patients with diabetes and patients with stroke. Ann Neurol. 2006;59:4–12.PubMedGoogle Scholar
- Winter DA, MacKinnon CD, Ruder GK, Wieman C. An integrated EMG/biomechanical model of upper body balance and posture during human gait. Prog Brain Res. 1993;97:359–67.PubMedGoogle Scholar
- Horak FB, Dozza M, Peterka R, Chiari L, Wall C. Vibrotactile biofeedback improves tandem gait in patients with unilateral vestibular loss. Ann N Y Acad Sci. 2009;1164:279–81.PubMed CentralPubMedGoogle Scholar
- Janssen LJF, Verhoeff LL, Horlings CGC, Allum JHJ. Directional effects of biofeedback on trunk sway during gait tasks in healthy young subjects. Gait Posture. 2009;29:575–81.PubMedGoogle Scholar
- Verhoeff LL, Horlings CGC, Janssen LJF, Bridenbaugh SA, Allum JHJ. Effects of biofeedback on trunk sway during dual tasking in the healthy young and elderly. Gait Posture. 2009;30:76–81.PubMedGoogle Scholar
- Sienko KH, Balkwill MD, Oddsson LIE, Wall C. The effect of vibrotactile feedback on postural sway during locomotor activities. J Neuroeng Rehabil. 2013;10:93.PubMed CentralPubMedGoogle Scholar
- Galica AM, Kang HG, Priplata AA, D’Andrea SE, Starobinets OV, Sorond FA, et al. Subsensory vibrations to the feet reduce gait variability in elderly fallers. Gait Posture. 2009;30:383–7.PubMed CentralPubMedGoogle Scholar
- Hewer RL. Rehabilitation after stroke. Q J Med. 1990;76:659–74.Google Scholar
- Huang H, Wolf SL, He J. Recent developments in biofeedback for neuromotor rehabilitation. J Neuroeng Rehabil. 2006;3:11.PubMed CentralPubMedGoogle Scholar
- Jonsdottir J, Cattaneo D, Recalcati M, Regola A, Rabuffetti M, Ferrarin M, et al. Task-oriented biofeedback to improve gait in individuals with chronic stroke: motor learning approach. Neurorehabil Neural Repair. 2010;24:478–85.PubMedGoogle Scholar
- Ding ZQ, Luo ZQ, Causo A, Chen IM, Yue KX, Yeo SH, et al. Inertia sensor-based guidance system for upperlimb posture correction. Med Eng Phys. 2013;35:269–76.PubMedGoogle Scholar
- Rao N, Aruin AS. Auxiliary sensory cues improve automatic postural responses in individuals with diabetic neuropathy. Neurorehabil Neural Repair. 2011;25:110–7.PubMedGoogle Scholar
- Redd CB, Bamberg SJM. A wireless sensory feedback device for real-time gait feedback and training. IEEE/ASME Trans Mechatronics. 2012;17:425–33.Google Scholar
- Van Wegen E, de Goede C, Lim I, Rietberg M, Nieuwboer A, Willems A, et al. The effect of rhythmic somatosensory cueing on gait in patients with Parkinson’s disease. J Neurol Sci. 2006;248:210–4.PubMedGoogle Scholar
- McKinney Z, Heberer K, Nowroozi BN, Greenberg M, Fowler E, Grundfest W: Pilot evaluation of wearable tactile biofeedback system for gait rehabilitation in peripheral neuropathy. IEEE Haptics Symp. Ieee; 2014:135–140Google Scholar
- Badke MB, Sherman J, Boyne P, Page S, Dunning K. Tongue-based biofeedback for balance in stroke: results of an 8-week pilot study. Arch Phys Med Rehabil. 2011;92:1364–70.PubMedGoogle Scholar
- Lee I, Choi S: Effects of Multi-modal Guidance for the Acquisition of Sight Reading Skills: A Case Study with Simple Drum Sequences. IEEE World Haptics; 2013:571–576.Google Scholar
- Spelmezan D, Jacobs M, Hilgers A, Borchers J: Tactile motion instructions for physical activities. Conf Hum Factors Comput Syst 2009:2243–2252.Google Scholar
- Dowling AV, Favre J, Andriacchi TP. Inertial sensor-based feedback can reduce key risk metrics for anterior cruciate ligament injury during jump landings. Am J Sports Med. 2012;40:1075–83.PubMedGoogle Scholar
- Wheeler JW, Shull PB, Besier T. Real-time knee adduction moment feedback for gait retraining through visual and tactile displays. J Biomech Eng. 2011;133:041007.PubMedGoogle Scholar
- Dowling AV, Fisher DS, Andriacchi TP. Gait modification via verbal instruction and an active feedback system to reduce peak knee adduction moment. J Biomech Eng. 2010;132:071007–5.PubMedGoogle Scholar
- Shull PB, Shultz R, Silder A, Dragoo JL, Besier TF, Cutkosky MR, et al. Toe-in gait reduces the first peak knee adduction moment in patients with medial compartment knee osteoarthritis. J Biomech. 2013;46:122–8.PubMedGoogle Scholar
- Shull PB, Silder A, Shultz R, Dragoo JL, Besier TF, Delp SL, et al. Six-week gait retraining program reduces knee adduction moment, reduces pain, and improves function for individuals with medial compartment knee osteoarthritis. J Orthop Res. 2013;31:1020–5.PubMedGoogle Scholar
- Lurie KL, Shull PB, Nesbitt KF, Cutkosky MR: Informing haptic feedback design for gait retraining. IEEE World Haptics; 2011:19–24.Google Scholar
- Jirattigalachote WJ, Shull PB, Cutkosky MR: Virtual pebble: a haptic state display for pedestrians. IEEE Ro-Man Symp; 2011:401–406.Google Scholar
- Alahakone AU, Senanayake SMNA: Vibrotactile feedback systems: Current trends in rehabilitation, sports and information display. IEEE/ASME Int Conf Adv Intell Mechatronics, AIM 2009:1148–1153.Google Scholar
- Rogers JA. A clear advance in soft actuators. Science. 2013;341:968–9.PubMedGoogle Scholar
- Majidi C. Soft robotics: A perspective—current trends and prospects for the future. Soft Robot. 2014;1:5–11.Google Scholar
- Yeo W-H, Kim Y-S, Lee J, Ameen A, Shi L, Li M, et al. Multifunctional epidermal electronics printed directly onto the skin. Adv Mater. 2013;25:2773–8.PubMedGoogle Scholar
- Riener R, Lünenburger L, Colombo G. Human-centered robotics applied to gait training and assessment. J Rehabil Res Dev. 2006;43:679.PubMedGoogle Scholar
- Xu S, Zhang Y, Cho J, Lee J, Huang X, Jia L, et al. Stretchable batteries with self-similar serpentine interconnects and integrated wireless recharging systems. Nat Commun. 2013;4:1543.PubMedGoogle Scholar
- Nyholm L, Nyström G, Mihranyan A, Strømme M. Toward flexible polymer and paper-based energy storage devices. Adv Mater. 2011;23:3751–69.PubMedGoogle Scholar
- Park S, Jayaraman S. Enhancing the quality of life through wearable technology. IEEE Eng Med Biol Mag. 2003;22:41–8.PubMedGoogle Scholar
- Axisa F, Schmitt PM, Gehin C, Delhomme G, Mcadams E, Dittmar A, et al. Flexible technologies and smart clothing for citizen medicine, home healthcare, and disease prevention. IEEE Trans Inf Technol Biomed. 2005;9:325–36.PubMedGoogle Scholar
- Poh M-Z, Swenson NC, Picard RW. A wearable sensor for unobtrusive, long-term assessment of electrodermal activity. IEEE Trans Biomed Eng. 2010;57:1243–52.PubMedGoogle Scholar
This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/4.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly credited. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.