- Open Access
Feasibility of the adaptive and automatic presentation of tasks (ADAPT) system for rehabilitation of upper extremity function post-stroke
© Choi et al; licensee BioMed Central Ltd. 2011
Received: 11 June 2010
Accepted: 3 August 2011
Published: 3 August 2011
Current guidelines for rehabilitation of arm and hand function after stroke recommend that motor training focus on realistic tasks that require reaching and manipulation and engage the patient intensively, actively, and adaptively. Here, we investigated the feasibility of a novel robotic task-practice system, ADAPT, designed in accordance with such guidelines. At each trial, ADAPT selects a functional task according to a training schedule and with difficulty based on previous performance. Once the task is selected, the robot picks up and presents the corresponding tool, simulates the dynamics of the tasks, and the patient interacts with the tool to perform the task.
Five participants with chronic stroke with mild to moderate impairments (> 9 months post-stroke; Fugl-Meyer arm score 49.2 ± 5.6) practiced four functional tasks (selected out of six in a pre-test) with ADAPT for about one and half hour and 144 trials in a pseudo-random schedule of 3-trial blocks per task.
No adverse events occurred and ADAPT successfully presented the six functional tasks without human intervention for a total of 900 trials. Qualitative analysis of trajectories showed that ADAPT simulated the desired task dynamics adequately, and participants reported good, although not excellent, task fidelity. During training, the adaptive difficulty algorithm progressively increased task difficulty leading towards an optimal challenge point based on performance; difficulty was then continuously adjusted to keep performance around the challenge point. Furthermore, the time to complete all trained tasks decreased significantly from pretest to one-hour post-test. Finally, post-training questionnaires demonstrated positive patient acceptance of ADAPT.
ADAPT successfully provided adaptive progressive training for multiple functional tasks based on participant's performance. Our encouraging results establish the feasibility of ADAPT; its efficacy will next be tested in a clinical trial.
Over 80% of first-time strokes (infarctions only) involve acute hemi-paresis of the upper limb . There is now definite evidence that intensive task-specific practice is effective in improving upper extremity function and use after stroke . In current clinical settings, however, at least in the US, patients with stroke receive only a small fraction of the dose of therapy needed for effective rehabilitation of upper extremity functions [3–6]. In view of this shortcoming of the conventional medical practice model, a growing number of robotic systems for the rehabilitation of upper extremities after stroke are being developed and tested [7–17]. Rehabilitation robots can potentially provide an automated and cost-effective addition to human therapists to increase the duration and intensity of therapy in rehabilitation settings.
Here, we argue that to further enhance the efficacy of therapy targeting the recovery of upper extremity functions, the design of rehabilitation robotic systems could benefit from three evidence-based rehabilitation principles: 1) training should maximize active participation from the patient; 2) training should involve practice of real-world tasks; and 3) training should be individualized by adaptively changing the number of trials and difficulty of each task practiced according to the evolving skill level of the patient.
Active participation training
The beneficial effects of self-produced movements over passive movements in the development of visually guided behavior have been long known . More recent work has shown that motor cortex reorganization underlies motor improvements during active motor training [19–22]. Although passive movements also elicit activity in the motor cortex, e.g. , actively generated movements are more effective in eliciting both performance improvements and cortical reorganization . Along these lines, training with active participation and repetitive practice has been shown to be effective in stroke rehabilitation . Recognizing that pure assistance may not be entirely beneficial for motor learning and recovery of arm and hand function [26, 27], rehabilitation roboticists have started to balance assistance provided by the robot with active movement by the patient, e.g. [28–30].
Task oriented training
Recovery of functional tasks constituting activities of daily life has been shown to be critical for patients in their attempts to return to a reasonable quality of life [31, 32] and improved participation . It is well established that the best way to learn an activity is to practice that activity, which means task-specific training , and there is mounting evidence that functional and meaningful task training is effective in stroke rehabilitation [35–38] (but see ). In particular, the manipulation of real and functional objects seems to be beneficial [40, 41]. As a result, Task-Oriented Training (TOT) has emerged as the dominant approach to motor restoration (NINDS Stroke Progress Review Group, 2006). To our knowledge, only two automated upper extremity rehabilitation systems allow functional task training with manipulation of real objects. First, AutoCite, which is a semi-automated (non robotic) system that allows patients to engage in the practice of functional tasks, has been shown to be as effective as standard constraint-induced (CI) therapy [42, 43]. AutoCite's hardware and software is capable of automatic task selection and difficulty adjustment based on previous performance patterns. However, in the preliminary testing with patients, no adaptive algorithm was tested, and task selection and adjustment of difficulty was done manually. Second, the robotic system ADLER presents multiple functional tasks for activities of daily living . Note however that ADLER requires a therapist to set up the different objects to be manipulated.
In motor learning, retention of skill over a prolonged period is strongly influenced by the number of trials and by the schedules in which the multiple tasks are practiced (e.g., random, blocked or interval-expanded presentation of tasks) [45–47]. Furthermore, challenging tasks, that is, tasks that are neither too difficult nor too easy, are most likely to elicit motor learning [19, 48–50]. Challenging tasks also enhance motivation, which may in turn further enhance learning . Because performance typically improves during skill acquisition according to negatively accelerated learning curves (but not always, see for instance ), and because learning evolves at different rates for each task and each subject, task difficulty needs to be dynamically adjusted to maintain challenge at an optimal level. During neuro-rehabilitation, it is likely that physical and occupational therapists adaptively modify task practice parameters using intuitive and largely implicit rules . Along these lines, a number of adaptive difficulty algorithms based on performance have been implemented on robotic systems [28, 54, 55].
In previous work, we developed a novel robotic system, the ADaptive and Automatic Presentation of Tasks (ADAPT) system, which implements all three rehabilitation principles mentioned above: ADAPT allows active and individualized training on a number of real-world functional tasks . ADAPT can accommodate an expanding number of such tasks, and it allows the implementation of performance-based adaptive task scheduling and task difficulty by simulating the dynamics of daily living functional tasks, such as opening a doorknob, opening a jar, turning a key, etc, each of which requires active reaching and manipulation.
The different tasks and tools were designed to engage training of different grasps (power, overhand, and pinch) and arm movements. Thus, like other automated rehabilitation devices, ADAPT allows both delivery of large intensity of training and longitudinal measurements of performance and limb use. However, the design of ADAPT departs from that of most other current robotic systems. In these systems, the task goal is generally pre-specified, and the robot modulates the assistance provided to the patient to attain that goal. Instead, ADAPT provides assistance-free training modeled after TOT. In this form of training, the rehabilitation therapist selects salient tasks for the patient to work on, adjusts the task parameters by increasing task difficulty and complexity as learning progresses, and provides informative and motivational feedback. Accordingly, in ADAPT, the task difficulty is constantly adapted such that patient always actively performs doable, but challenging, tasks.
The primary aim of this feasibility study was to establish the feasibility of ADAPT for a single training session of participants with chronic stroke. Specifically, we evaluated ADAPT's safety, ADAPT's overall functionality, possible improvement of performance between pre- and post-test, and participants' subjective experience.
Safety was measured both quantitatively by the number of adverse event occurring in the operation of the ADAPT and qualitatively via a participant questionnaire similar to that used in other robot acceptance studies such as in [57, 58].
Functionality, which is broadly defined as the robot applicability toward the accomplishment of a task , was evaluated in three ways. First, we tested whether ADAPT could successfully present the different tasks to the participants without human intervention. Second, we evaluated the fidelity of the dynamics of the simulated tasks both by comparing it to actual task dynamics and via questionnaire. Third, we evaluated whether the adaptive algorithm could successfully modulate task difficulty based on performance during training.
Improvement in performance was measured by the time it takes the subject to perform the 6 different tasks between pre- and post-test.
We limited our target population to individuals with chronic stroke who have some residual arm and hand movements; that is, participants in our study can be classified as "moderately to mildly" impaired . Such participants were included for two reasons. First, because the safety of ADAPT had only been tested with healthy participants, we included participants who had the potential to easily disengage themselves from the task whether an unexpected robot behavior would occur. Second, because ADAPT does not provide any physical assistance, but instead require active engagement to perform reach, grasp, and manipulate objects, it is designed for patients with residual volitional motor capability of the arm and hand.
Participants characteristics. Hand dominance is before stroke
Fugl-meyer score (0-66)
The ADAPT robotic system
List of Functional Tasks
Turn a door knob with power grasping to the end of turning range, and release it.
Push a door bell with a finger over a threshold force.
Turn a jar with power grasping up to the end of turning range.
Move a door knob horizontally with power grasping. Mostly elbow and shoulder motion.
Turn a door knob button with pinch grasping to the end of turning range.
Turn a screwdriver with power grasping to the end of turning range.
where τ is the torque exerted by a subject, Diff is an adjustable difficulty variable, q, , are angle, velocity, acceleration respectively of task motion, and is the dynamics model of an original functional task. For three tasks (doorknob turning, jar opening and door-locking tasks), we modeled the functional task dynamics using a constructive locally weighted method, such that non-linear dynamics (like friction) are accurately modeled to simulate the feel of task tools . This approach provides computationally efficient way of modeling non-linear dynamics via the combination of multiple local linear models by adaptively creating and pruning the local models (see  for details). For bell pushing and door opening, we used simple mathematical models;for screw driving we reused the jar opening model captured by our original model by changing the parameters of the local linear models empirically.
where Difft, k is the difficulty for the current trial t and the task k, α is learning rate, Perft-1, k is the performance on the previous trial, and Perfref, k is the reference challenging performance. Task difficulty is updated to constantly maintain performance around the "challenging point", Perfref, k (see  for a detail of this algorithm).
The low-level admittance controller computes the control current corresponding to the desired trajectory to implement the desired dynamics of the task with difficulty prescribed by the high-level controller. The computed control current is then applied to the robot to simulate the task, and the patient feels the simulated dynamics by manipulating the selected tool (see  for details).
Safety was a crucial issue in the design of ADAPT. Our choice of design makes our robot safer than a traditional multi-DOF robotic arm because of the small overall workspace. The linear DOF is only used for tool positioning, not for task dynamics simulation. For simplicity and safety reasons, we choose tasks that require movements around a single DOF during subject-robot interactions. After a task is set up for presentation to the subject, the magnetic brakes that are built into the robotic articulations are engaged on the other three DOFs during subject-robot interactions. Furthermore, the patient is not strapped to the robot but freely interact with the robot, only after the robot positions the tool.
Several surveillance routines are implemented to limit the maximal torque output and cap the maximum velocity of the linear and rotational motors. Watchdog routines that continuously check for failure of the position and force sensors, computer crashes, and electrical failures automatically freeze the robot by engaging the magnetic breaks in all DOFs. When pressed, two emergency stop buttons stop all robot operation and turn on magnetic brakes to disable any movement of all 4 DOF of the robot. The main emergency red stop button of the power box in is accessible to the therapist. The subject holds the second emergency stop button at all times with their less affected hand.
Participants sat on a chair in front of ADAPT and were instructed, at each trial, to reach and manipulate a functional task tool, which was selected by the adaptive scheduler and then picked up and presented by the robot. The experiment started with a pre-test followed by a training session, and ended with a post-test after one hour break. In the pre-test and post-test, participants practiced all six tasks without feedback with difficulty set at the easiest levels, in block of three trials each (total 18 trials). Then, four tasks were selected from the six tasks of the pre-test for the training session. If a participant could not complete a specific task in the pre-test, that task was not selected. Otherwise, the four tasks were selected pseudo-randomly, so as to counterbalance the number of tasks across participants. The training session, which lasted about one and half hour, consisted of 48 blocks of three trials (12 blocks per task), pseudo-randomly scheduled so that each task occurred once in any four consecutive blocks.
Fidelity is the extent to which the appearance and behavior of the simulator/simulation matches the appearance and behavior of the simulated system. We tested the two components of fidelity: engineering fidelity, which is 'the degree to which the simulator replicates the physical characteristics of the real task', and psychological or functional fidelity, which is 'the degree to which the skill or skills in the real task are captured in the simulated task'. We studied engineering fidelity by checking whether the tasks have the qualitative mechanical characteristics of the tasks modeled. We quantified psychological fidelity via the participant questionnaire.
The participants practiced the selected four tasks in the adaptive difficulty mode, in which the difficulty of each task was updated according to equation 1. Gradual and progressive motor adaptation promotes longer-lasting effect in motor learning compare to sudden adaptation . Therefore, the task difficulty was set to be lowest initially, and increased progressively based on performance. Additionally, this set up led to high probability of success in initial trials, which is presumably important for high motivation . Due to a high learning rate in equation (1), task difficulty increased rapidly, so that performance reached the challenging point before the end of training in most cases. We recorded the number of trials need for the difficulty level to stop increasing and then, for the remaining trials, the deviation from the challenge point.
We assessed improvements in performance due to training by comparing median movement times (over three trials) on the tasks in the pre- and post-tests. Because all six tasks were given in the pre- and post-tests, but only four tasks were practiced during training, there were 20 out of 30 (five participants by six tasks) trained tasks, and 10 untrained tasks. We performed paired sample t-tests on the median movement times to analyze the effect of training. We report results as means ± standard deviations. Our significance level was p < .05.
Following the post-test, two questionnaires were administered to evaluate the acceptance of ADAPT. We first administered the Intrinsic Motivation Inventory (IMI) questionnaire, which was designed to assess participants' subjective experience related to a target activity in laboratory experiments [51, 60], and has been previously used to measure stroke patients' experience in robotic training . Second, we administered a specific questionnaire, similar to those used in other robot acceptance studies such as in [57, 58], to inquire about perceived safety, fidelity of simulated tasks and subjective experience.
All five participants completed the robotic training sessions, functional measurements, and questionnaires. Safety concerns were strongly addressed from the initial design process of ADAPT (see above and ), and no adverse event occurred during direct interaction between participants and ADAPT. In particular, the two safety buttons were never activated. Finally patient reported very high sense of safety in the post-training questionnaire (6.80 ± 0.45 out of 7).
Automatic presentations of functional tasks without human intervention
ADAPT could successfully present the six tasks to all 5 participants without human intervention. The robot thus implemented a total of 900 trials (18 pre-test trials + 48 * 3 training trials + 18 post-test trials for 5 subjects) without any failure either in the robot itself or in the pneumatic tool changer.
Fidelity of Functional Task
Participants reported that they could clearly understand how to interact with ADAPT through auditory and visual instructions, and that the simulated functional tasks were fairly similar to real tasks (participants scored 5.90 ± 1.6 out of 7 when asked if the tasks were similar to real functional tasks). One participant verbally reported that pushing bell required too much force, suggesting that tasks with high difficulty could be unrealistic.
Maintaining challenging performance via adaptive difficulty
We then analyzed the performance of the adaptive algorithms for each task across participants. Since the difficulty level was set at the easiest level for all tasks and participants, it took an average of 29.4 ± 8.2 trials for the difficulty level to stop increasing. Then for the remaining trials, difficulty stayed within 33.3 ± 27.9% of the challenge point.
Improvements in movement time
Acceptance of ADAPT
Subscale findings of the IMI questionnaire administered after training (subscale range = 1 - 7)
Score (Mean ± SD)
6.17 ± 1.25
4.90 ± 1.88
6.32 ± 0.75
6.34 ± 1.43
1.84 ± 0.94
6.43 ± 1.40
Patients' acceptance for the training session with ADAPT (subscale range = 1 - 7)
Score (Mean ± SD)
Comfortable with robot's sound & appearance
6.90 ± 0.32
6.80 ± 0.45
Comfortable with seatbelt & chair
6.10 ± 1.52
Similar to real functional tasks
5.90 ± 1.62
Fatigue or Frustration
3.40 ± 1.95
7.00 ± 0.00
In this study, we tested and evaluated the feasibility of ADAPT with five participants with chronic stroke during one day training session. Specifically, we evaluated safety, overall functionality, fidelity of simulated tasks, improvement in performance, and patient acceptance.
Safety was excellent since no adverse event occurred during training, and the emergency stop buttons were needed by neither the participants nor the experimenter. Results from questionnaire showed excellent perceived safety. Note that the robot exerts large movements when picking up new objects, but such robot movements were programmed at low speed. We also fixed the work table between the robot and the participants such that accidental contact between the robot and the participant during this phase is unlikely.
For all participants and all tasks studied, ADAPT could accomplish the following functions satisfactorily. 1. Present a variety of functional tasks without human intervention 2. Simulate the dynamics of these functional tasks and 3. Modulate adaptive task difficulty based on performance during training,
First ADAPT could present a variety of functional tasks without human intervention for a total of 900 trials. This shows the feasibility of our novel tool changing system, based on a pneumatic tool changer, which automatically selected one of four tools based on the task schedule. No failure with the tool changing process occurred for all five participants' testing and training sessions. Extension to a greater number of tools in future work is straightforward.
Second, ADAPT could simulate the dynamics of the functional tasks qualitatively, as demonstrated by showing that the actual tasks had the characteristics of the simulated tasks for all six tasks. In addition, participants reported good, although not excellent, perceived fidelity. Two reasons may have led to this result. First, we modeled the functional task dynamics with constructive locally weighted method only for doorknob turning, jar opening and door-locking. In contrast, we used simple mathematical models for bell pushing and door opening, and we reused the jar opening model for screw driving by changing the parameters of the local linear models empirically. Second, we required large, and increasingly large, amount of force for most tasks. The difficulty of current tasks was mostly proportional to torque or force to be exerted for manipulation. Although strength is beneficial for stroke recovery [37, 64], many functional tasks do not require much motor strength, but instead emphasize skill in fine motor coordination. The strength-oriented difficulty algorithm that we used here may cause a task to be perceived as more difficult than the same task in the real world. In future work, we will need to develop tasks that require fine motor skills and derive algorithms that modulate difficulty based on measures of skill of performance such as movement time and/or errors, not strength.
Third, testing of adaptive modulation of task difficulty based on performance during training showed that the adaptive algorithm could reasonably well adapt difficulty to the participant's performance following the initial increase in performance, as performance stayed within 33% of the challenge performance, on average. Such deviation from the challenge performance can occur both as a result of participants behavior (trial-by-trial noise and actual change in performance, due to learning or fatigue) and as a result of the form of our adaptive difficulty updating algorithm (equation 2) and its learning rate parameter α, which was set based on data from healthy participants. If the learning rate is too high, the difficulty will follow the noise in the data and large oscillations may occur (possibly as observed in Figure 5b); if too low, the difficulty will lag performance. Additional computer simulations and experiments are needed to define adaptive rules that reduce the deviation between desired and actual performance and capturing actual slow changing changes in participants' performance. Similarly, the parameters for challenging performance Perf ref,k were also set empirically for each task based on pilot data with healthy subjects. Indeed, how to determine the challenge point parameters systematically is an open question ; in addition, a challenging performance parameter that exceeds real-life values may not have much functional relevance. The final limitation of our adaptive difficulty algorithm is that comparison of performance between different tasks is difficult, because the performance value is a physical quantity, such as rotated angle or pushing force. Using success rate as a performance metric is a possible solution to this problem, and efforts are underway to develop statistical models of performance for each task and each subject.
We showed that ADAPT could improve a measure of participants' performance, time to movement time of the task trained, in a single session. Note that while we previously showed that adaptive difficulty can presumably outperform fixed difficulty in motor learning , the current study was not designed to specifically test whether the adaptive difficulty schedule was effective. Thus, the gains in movement time may however be unrelated to the adaptive algorithm, as the practice itself could have provided the reduction in movement time. Furthermore, although the current ADAPT system allows to implement adaptive schedule in the number of trials, such as that we previously developed in , we implemented here a simple pseudo-random schedule in which the trial numbers were equal for all tasks. Our current adaptive task scheduler requires at least a first "fixed session" to determine the number of trials on subsequent sessions based on performance on delayed retention . Here, because safety with stroke participants was not established before this study, we conducted a single session and thus fixed the number of trials. Furthermore, because the retention test was given 1 hour after practice, our results showing improvement in movement time may be due to short-term practice effects and not true learning. In further work, training for more than one session, and a longer post time measurement will be needed to extrapolate to a long-term benefit for stroke patients. Therefore, in its current form, and because we show no transfer to other tasks (even similar ones), a single session with ADAPT should not be viewed as a comprehensive rehabilitation program, but rather a possible automated module within a rehabilitation program.
Because ADAPT in its current configuration does not constrain the types of reaching and grasping movements performed, patients may use compensatory movements to accomplish the task goal. Videos showed that indeed several patients did not manipulate the task tools as instructed, even though the participants knew the appropriate way of manipulation. In the door locking task for instance, which requires pinch grasping, one participant (FM score: 42) used power grasping to succeed in the trial. Thus, on one hand, compensatory movements could be reduced with tools that constrain hand or finger postures and motion monitoring systems. On the other hand, it has been argued that stroke recovery therapy should not focus solely on the realization of normal movement patterns, but should take some compensatory strategy into account to be more effective .
The results from this study validate the feasibility of ADAPT for rehabilitation of arm and hand function after stroke, and provide justification for continued investigation of clinical efficacy. Furthermore, safe automatic presentation of functional tasks with ADAPT showed the potential to engage effective motor learning in stroke rehabilitation. Motor control and learning principles, such as manipulating the schedule of training, can extend the efficacy of robotic neuro-rehabilitation . This motivates our current and future work in developing hypothesis-driven adaptive schedules to select the appropriate task and difficulty at each trial for optimal stroke rehabilitation.
This work was supported in part in by the Division of Biokinesiology and Physical Therapy, University of Southern California, and in part by the National Institutes of Health under Grant R03 HD050591-02. The authors thank Hui-ting Goh and Shuya Chen for their help in participant recruitment and during the experiment, as well as Carolee Winstein for helpful discussions.
- Desrosiers J, Malouin F, Bourbonnais D, Richards CL, Rochette A, Bravo G: Arm and leg impairments and disabilities after stroke rehabilitation: relation to handicap. Clin Rehabil 2003,17(6):666-73. 10.1191/0269215503cr662oaView ArticlePubMedGoogle Scholar
- Wolf SL, Winstein CJ, Miller JP, Taub E, Uswatte G, Morris D, Giuliani C, Light KE, Nichols-Larsen D: Effect of constraint-induced movement therapy on upper extremity function 3 to 9 months after stroke: the EXCITE randomized clinical trial. Jama 2006,296(17):2095-104. 10.1001/jama.296.17.2095View ArticlePubMedGoogle Scholar
- Lincoln NB, Willis D, Philips SA, Juby LC, Berman P: Comparison of rehabilitation practice on hospital wards for stroke patients. Stroke 1996,27(1):18-23. 10.1161/01.STR.27.1.18View ArticlePubMedGoogle Scholar
- Keith RA, Cowell KS: Time use of stroke patients in three rehabilitation hospitals. Soc Sci Med 1987,24(6):529-33. 10.1016/0277-9536(87)90342-XView ArticlePubMedGoogle Scholar
- Mackey F, Ada L, Heard R, Adams R: Stroke rehabilitation: are highly structured units more conducive to physical activity than less structured units? Arch Phys Med Rehabil 1996,77(10):1066-70. 10.1016/S0003-9993(96)90070-2View ArticlePubMedGoogle Scholar
- Lang C, Macdonald J, Reisman D, Boyd L, Kimberley Jacobson T, Schindler-Ivens S, Hornby T, Ross S, Scheets P: Observation of amounts of movement practice provided during stroke rehabilitation. Arch Phys Med Rehabil 2009,90(10):1692-8. 10.1016/j.apmr.2009.04.005PubMed CentralView ArticlePubMedGoogle Scholar
- Aisen ML, Krebs HI, Hogan N, McDowell F, Volpe BT: The effect of robot-assisted therapy and rehabilitative training on motor recovery following stroke. Arch Neurol 1997,54(4):443-6.View ArticlePubMedGoogle Scholar
- Burgar CG, Lum PS, Shor PC, Van der Loos Machiel HF: Development of robots for rehabilitation therapy: the Palo Alto VA/Stanford experience. J Rehabil Res Dev 2000,37(6):663-73.PubMedGoogle Scholar
- Reinkensmeyer DJ, Kahn LE, Averbuch M, McKenna-Cole A, Schmit BD, Rymer WZ: Understanding and treating arm movement impairment after chronic brain injury: progress with the ARM guide. J Rehabil Res Dev 2000,37(6):653-62.PubMedGoogle Scholar
- Hesse S, Schulte-Tigges G, Konrad M, Bardeleben A, Werner C: Robot-assisted arm trainer for the passive and active practice of bilateral forearm and wrist movements in hemiparetic subjects. Arch Phys Med Rehabil 2003,84(:6):915-20.View ArticlePubMedGoogle Scholar
- Krebs HI, Volpe BT, Williams D, Celestino J, Charles SK, Lynch D, Hogan N: Robot-aided neurorehabilitation: A robot for wrist rehabilitation. Ieee Transactions on Neural Systems and Rehabilitation Engineering 2007,15(3):327-335.PubMed CentralView ArticlePubMedGoogle Scholar
- Dovat L, Lambercy O, Gassert R, Maeder T, Milner T, Leong TC, Burdet E: HandCARE: a cable-actuated rehabilitation system to train hand function after stroke. IEEE Trans Neural Syst Rehabil Eng 2008,16(6):582-91.View ArticlePubMedGoogle Scholar
- Krebs HI, Aisen ML, Volpe BT, Hogan N: Quantization of continuous arm movements in humans with brain injury. Proc Natl Acad Sci USA 1999,96(8):4645-9. 10.1073/pnas.96.8.4645PubMed CentralView ArticlePubMedGoogle Scholar
- Krebs HI, Hogan N, Volpe BT, Aisen ML, Edelstein L, Diels C: Overview of clinical trials with MIT-MANUS: a robot-aided neuro-rehabilitation facility. Technol Health Care 1999,7(6):419-23.PubMedGoogle Scholar
- Lum PS, Burgar CG, Shor PC, Majmundar M, Van der Loos M: Robot-assisted movement training compared with conventional therapy techniques for the rehabilitation of upper-limb motor function after stroke. Arch Phys Med Rehabil 2002,83(7):952-9. 10.1053/apmr.2001.33101View ArticlePubMedGoogle Scholar
- Lo AC, Guarino PD, Richards LG, Haselkorn JK, Wittenberg GF, Federman DG, Ringer RJ, Wagner TH, Krebs HI, Volpe BT, Bever CT Jr, Bravata DM, Duncan PW, Corn BH, Maffucci AD, Nadeau SE, Conroy SS, Powell JM, Huang GD, Peduzzi P: Robot-assisted therapy for long-term upper-limb impairment after stroke. N Engl J Med 2010,362(19):1772-83. 10.1056/NEJMoa0911341View ArticlePubMedGoogle Scholar
- Alamri A, Jongeun C, Saddik El A: AR-REHAB: An Augmented Reality Framework for Poststroke-Patient Rehabilitation. Instrumentation and Measurement, IEEE Transactions on 2010,59(10):2554-2563.View ArticleGoogle Scholar
- Held R, Hein A: Movement-Produced Stimulation in the Development of Visually Guided Behavior. J Comp Physiol Psychol 1963, 56: 872-6.View ArticlePubMedGoogle Scholar
- Nudo RJ, Wise BM, SiFuentes F, Milliken GW: Neural substrates for the effects of rehabilitative training on motor recovery after ischemic infarct. Science 1996,272(5269):1791-4. 10.1126/science.272.5269.1791View ArticlePubMedGoogle Scholar
- Karni A: When practice makes perfect. Lancet 1995,345(8946):395.View ArticlePubMedGoogle Scholar
- Muellbacher W, Ziemann U, Boroojerdi B, Cohen L, Hallett M: Role of the human motor cortex in rapid motor learning. Exp Brain Res 2001,136(4):431-8. 10.1007/s002210000614View ArticlePubMedGoogle Scholar
- Conner JM, Culberson A, Packowski C, Chiba AA, Tuszynski MH: Lesions of the Basal forebrain cholinergic system impair task acquisition and abolish cortical plasticity associated with motor skill learning. Neuron 2003,38(5):819-29. 10.1016/S0896-6273(03)00288-5View ArticlePubMedGoogle Scholar
- Carel C, Loubinoux I, Boulanouar K, Manelfe C, Rascol O, Celsis P, Chollet F: Neural substrate for the effects of passive training on sensorimotor cortical representation: a study with functional magnetic resonance imaging in healthy subjects. J Cereb Blood Flow Metab 2000,20(3):478-84.View ArticlePubMedGoogle Scholar
- Lotze M, Braun C, Birbaumer N, Anders S, Cohen LG: Motor learning elicited by voluntary drive. Brain 2003,126(Pt 4):866-72.View ArticlePubMedGoogle Scholar
- Duncan PW: Synthesis of intervention trials to improve motor recovery following stroke. Tpoics in Stroke Rehabilitation 1997, 3: 1-20.Google Scholar
- Marchal-Crespo L, Reinkensmeyer DJ: Review of control strategies for robotic movement training after neurologic injury. J Neuroeng Rehabil 2009, 6: 20. 10.1186/1743-0003-6-20PubMed CentralView ArticlePubMedGoogle Scholar
- Kahn LE, Lum PS, Rymer WZ, Reinkensmeyer DJ: Robot-assisted movement training for the stroke-impaired arm: Does it matter what the robot does? J Rehabil Res Dev 2006,43(5):619-30. 10.1682/JRRD.2005.03.0056View ArticlePubMedGoogle Scholar
- Krebs H, Palazzolo J, Dipietro L, Volpe B, Hogan N: Rehabilitation robotics: Performance-based progressive robot-assisted therapy. AUTONOMOUS ROBOTS 2003,15(1):7-20. 10.1023/A:1024494031121View ArticleGoogle Scholar
- Kahn LE, Rymer WZ, Reinkensmeyer DJ: Adaptive assistance for guided force training in chronic stroke. In 26th Annual International Conference of the IEEE EMBS. San Francisco, CA; 2004.Google Scholar
- Dipietro L, Ferraro M, Palazzolo JJ, Krebs HI, Volpe BT, Hogan N: Customized interactive robotic treatment for stroke: EMG-triggered therapy. IEEE Trans Neural Syst Rehabil Eng 2005,13(3):325-34. 10.1109/TNSRE.2005.850423PubMed CentralView ArticlePubMedGoogle Scholar
- Ahlsio B, Britton M, Murray V, Theorell T: Disablement and Quality of Life after Stroke. Stroke 1984,15(5):886-890. 10.1161/01.STR.15.5.886View ArticlePubMedGoogle Scholar
- Carr JH, Shepherd RB: Stroke Rehabilitation. Guidelines for Exercise and Training to Optimise Motor Skill. Oxford: Butterworth Heinemann; 2003.Google Scholar
- Duncan PW, Wallace D, Lai SM, Johnson D, Embretson S, Laster LJ: The stroke impact scale version 2.0. Evaluation of reliability, validity, and sensitivity to change. Stroke 1999,30(10):2131-40. 10.1161/01.STR.30.10.2131View ArticlePubMedGoogle Scholar
- Richards CL, Malouin F, Wood-Dauphinee S, Williams JI, Bouchard JP, Brunet D: Task-specific physical therapy for optimization of gait recovery in acute stroke patients. Arch Phys Med Rehabil 1993,74(6):612-20. 10.1016/0003-9993(93)90159-8View ArticlePubMedGoogle Scholar
- Fisher BE, Sullivan KJ: Activity-dependent factors affecting poststroke functional outcomes. Top Stroke Rehabil 2001,8(3):31-44. 10.1310/B3JD-NML4-V1FB-5YHGView ArticlePubMedGoogle Scholar
- Van Peppen RP, Kwakkel G, Wood-Dauphinee S, Hendriks HJ, Van der Wees PJ, Dekker J: The impact of physical therapy on functional outcomes after stroke: what's the evidence. Clin Rehabil 2004, 18: 833-62. 10.1191/0269215504cr843oaView ArticlePubMedGoogle Scholar
- Winstein CJ, Rose DK, Tan SM, Lewthwaite R, Chui HC, Azen SP: A randomized controlled comparison of upper-extremity rehabilitation strategies in acute stroke: A pilot study of immediate and long-term outcomes. Arch Phys Med Rehabil 2004,85(4):620-8. 10.1016/j.apmr.2003.06.027View ArticlePubMedGoogle Scholar
- Wolf SL, Thompson PA, Morris DM, Rose DK, Winstein CJ, Taub E, Giuliani C, Pearson SL: The EXCITE trial: attributes of the Wolf Motor Function Test in patients with subacute stroke. Neurorehabil Neural Repair 2005,19(3):194-205. 10.1177/1545968305276663View ArticlePubMedGoogle Scholar
- Krebs H, Mernoff S, Fasoli S, Hughes R, Stein J, Hogan N: A comparison of functional and impairment-based robotic training in severe to moderate chronic stroke: A pilot study. NeuroRehabilitation 2008, 23: 81-87.PubMedGoogle Scholar
- van Vliet P, Kerwin DG, Sheridan MR, Fentem P: The influence of goals on the kinematics of reaching forllowing stroke. Neurology Report 1995, 19: 11-16.View ArticleGoogle Scholar
- Wu C, Trombly CA, Lin K, Tickle-Degnen L: A kinematic study of contextual effects on reaching performance in persons with and without stroke: influences of object availability. Arch Phys Med Rehabil 2000,81(1):95-101.View ArticlePubMedGoogle Scholar
- Lum PS, Taub E, Schwandt D, Postman M, Hardin P, Uswatte G: Automated Constraint-Induced Therapy Extension (AutoCITE) for movement deficits after stroke. J Rehabil Res Dev 2004,41(3A):249-58. 10.1682/JRRD.2003.06.0092View ArticlePubMedGoogle Scholar
- Taub E, Lum PS, Hardin P, Mark VW, Uswatte G: AutoCITE: automated delivery of CI therapy with reduced effort by therapists. Stroke 2005,36(6):1301-4. 10.1161/01.STR.0000166043.27545.e8View ArticlePubMedGoogle Scholar
- Johnson MJ, Wisneski KJ, J A, Nathan D, S R: Development of ADLER:The Activities of Daily Living Exercise Robot. In IEEE-EMBS Biomedical Robotics (BioRob 2006). Pisa, Italy; 2006.Google Scholar
- Choi Y, Qi F, Gordon J, Schweighofer N: Performance-based adaptive schedules enhance motor learning. J Mot Behav 2008,40(4):273-80. 10.3200/JMBR.40.4.273-280View ArticlePubMedGoogle Scholar
- Goh H, Choi YG, Stewart JC, Lewthwaite R, Winstein CJ, Schweighofer N: Grip force modulation after stroke: what is the optimal schedule? In WCPT. Vancouver, Canada; 2007.Google Scholar
- Magill RA, Hall KG: A review of the contextual interference effect in motor skill acquisition. Human Movement Science 9th edition. 1990, 241-289.Google Scholar
- Plautz E, Milliken GW, Nudo RJ: Effects of repetitive motor training on movement representations in adult squirrel monkeys: role of use versus learning. Neurobiol Learn Mem 2000, 74: 27-55. 10.1006/nlme.1999.3934View ArticlePubMedGoogle Scholar
- Sanger TD: Failure of motor learning for large initial errors. Neural Comput 2004,16(9):1873-86. 10.1162/0899766041336431View ArticlePubMedGoogle Scholar
- Vygotsky LS: Mind in Society. Cambridge, MA: Harvard University Press; 1978.Google Scholar
- Deci E, Ryan RM: Intrinsic motivation and self-determination in human behavior. New York: Plenum; 1985.View ArticleGoogle Scholar
- Brooks V, Hilperath F, Brooks M, Ross H, Freund H: Learning what and how in a human motor task. Learn Mem 1995, 2: 225-42. 10.1101/lm.2.5.225View ArticlePubMedGoogle Scholar
- Gordon J: Assumptions underlying physical therapy intervention: Theoretical and historical perspectives. In Movement science: Foundations for physical therapy in rehabilitation. 2nd edition. Edited by: Carr JH, Shepherd RB. Aspen Publishers: Rockville, MD; 2000:1-31.Google Scholar
- Erlandson RF, deBear P, Kristy K, Dijkers M, Wu S: A robotic system to provide movement therapy. In Proc 5th Int Service Robot Conf. Detroit, MI; 1990.Google Scholar
- Emken JL, Benitez R, Sideris A, Bobrow JE, Reinkensmeyer DJ: Motor adaptation as a greedy optimization of error and effort. J Neurophysiol 2007,97(6):3997-4006. 10.1152/jn.01095.2006View ArticlePubMedGoogle Scholar
- Choi Y, Gordon J, Kim D, Schweighofer N: An Adaptive Automated Robotic Task-Practice System for Rehabilitation of Arm Functions After Stroke. IEEE Trans in Robotics 2009,25(3):556-568.View ArticleGoogle Scholar
- Krebs H, Hogan N, Aisen M, Volpe B: Robot-aided neurorehabilitation. IEEE Trans Rehabil Eng 1998,6(1):75-87. 10.1109/86.662623PubMed CentralView ArticlePubMedGoogle Scholar
- Dijkers M, deBear P, Erlandson R, Kristy K, Geer D, Nichols A: Patient and staff acceptance of robotic technology in occupational therapy: a pilot study. J Rehabil Res Dev 1991,28(2):33-44. 10.1682/JRRD.1991.04.0033View ArticlePubMedGoogle Scholar
- Bogoni L, Bajcsy R: An Active Approach to Characterization and Recognition of Functionality and Functional Properties. Department of Computer & Information Science, University of Pennsylvania; 1993.Google Scholar
- Ryan R, Deci E: Self-determination theory and the facilitation of intrinsic motivation, social development, and well-being. Am Psychol 2000,55(1):68-78.View ArticlePubMedGoogle Scholar
- Colombo R, Pisano F, Mazzone A, Delconte C, Micera S, Carrozza M, Dario P, Minuco G: Design strategies to improve patient motivation during robot-aided rehabilitation. J Neuroeng Rehabil 2007, 4: 3. 10.1186/1743-0003-4-3PubMed CentralView ArticlePubMedGoogle Scholar
- Worsnopp TT, Peshkin MA, Colgate JE, Kamper DG: An Actuated Finger Exoskeleton for Hand Rehabilitation Following Stroke. In IEEE 10th International Conference on Rehabilitation Robotics. Noordwijk, The Netherlands; 2007.Google Scholar
- Huang V, Krakauer J: Robotic neurorehabilitation: a computational motor learning perspective. J Neuroeng Rehabil 2009, 6: 5. 10.1186/1743-0003-6-5PubMed CentralView ArticlePubMedGoogle Scholar
- Ada L, Dorsch S, Canning C: Strengthening interventions increase strength and improve activity after stroke: a systematic review. Aust J Physiother 2006,52(4):241-8. 10.1016/S0004-9514(06)70003-4View ArticlePubMedGoogle Scholar
- Guadagnoli MA, Lee TD: Challenge point: a framework for conceptualizing the effects of various practice conditions in motor learning. J Mot Behav 2004,36(2):212-24. 10.3200/JMBR.36.2.212-224View ArticlePubMedGoogle Scholar
- Kwakkel G: Towards integrative neurorehabilitation science. Physiother Res Int 2009,14(3):137-46. 10.1002/pri.446View ArticlePubMedGoogle Scholar
- Krakauer JW: Motor learning: its relevance to stroke recovery and neurorehabilitation. Curr Opin Neurol 2006,19(1):84-90. 10.1097/01.wco.0000200544.29915.ccView ArticlePubMedGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.