Development of a multi-electrode array for spinal cord epidural stimulation to facilitate stepping and standing after a complete spinal cord injury in adult rats
- Parag Gad†1, 3,
- Jaehoon Choe†2, 3,
- Mandheerej Singh Nandra†7,
- Hui Zhong†3,
- Roland R Roy3, 6,
- Yu-Chong Tai†7, 8, 9 and
- V Reggie Edgerton†3, 4, 5Email author
© Gad et al; licensee BioMed Central Ltd. 2013
Received: 15 February 2012
Accepted: 7 January 2013
Published: 21 January 2013
The Erratum to this article has been published in Journal of NeuroEngineering and Rehabilitation 2015 12:33
Stimulation of the spinal cord has been shown to have great potential for improving function after motor deficits caused by injury or pathological conditions. Using a wide range of animal models, many studies have shown that stimulation applied to the neural networks intrinsic to the spinal cord can result in a dramatic improvement of motor ability, even allowing an animal to step and stand after a complete spinal cord transection. Clinical use of this technology, however, has been slow to develop due to the invasive nature of the implantation procedures, the lack of versatility in conventional stimulation technology, and the difficulty of ascertaining specific sites of stimulation that would provide optimal amelioration of the motor deficits. Moreover, the development of tools available to control precise stimulation chronically via biocompatible electrodes has been limited. In this paper, we outline the development of this technology and its use in the spinal rat model, demonstrating the ability to identify and stimulate specific sites of the spinal cord to produce discrete motor behaviors in spinal rats using this array.
We have designed a chronically implantable, rapidly switchable, high-density platinum based multi-electrode array that can be used to stimulate at 1–100 Hz and 1–10 V in both monopolar and bipolar configurations to examine the electrophysiological and behavioral effects of spinal cord epidural stimulation in complete spinal cord transected rats.
In this paper, we have demonstrated the effectiveness of using high-resolution stimulation parameters in the context of improving motor recovery after a spinal cord injury. We observed that rats whose hindlimbs were paralyzed can stand and step when specific sets of electrodes of the array are stimulated tonically (40 Hz). Distinct patterns of stepping and standing were produced by stimulation of different combinations of electrodes on the array located at specific spinal cord levels and by specific stimulation parameters, i.e., stimulation frequency and intensity, and cathode/anode orientation. The array also was used to assess functional connectivity between the cord dorsum to interneuronal circuits and specific motor pools via evoked potentials induced at 1 Hz stimulation in the absence of any anesthesia.
Therefore the high density electrode array allows high spatial resolution and the ability to selectively activate different neural pathways within the lumbosacral region of the spinal cord to facilitate standing and stepping in adult spinal rats and provides the capability to evoke motor potentials and thus a means for assessing connectivity between sensory circuits and specific motor pools and muscles.
It is well established that the spinal cord contains intricate computing units capable of performing rapid ongoing motor processing of complex proprioceptive and cutaneous input during coordinated motor behaviors such as standing and stepping. Neural networks in the lumbosacral spinal cord (i.e., central pattern generators (CPG)) can function autonomously (without any brain control) to produce the characteristic alternating motor patterns of gait and to compensate for errors and obstacles[2, 3] using only sensory information from the limbs[4–7]. More recently it has become recognized that these networks have the ability to process complex sensory ensembles that can serve as the controller of posture and locomotion[6, 8, 9].
The rat or cat spinal cord isolated from supraspinal control via a complete low- to mid-thoracic spinal cord transection produces locomotor-like patterns in the hindlimbs when facilitated pharmacologically and/or by epidural spinal cord stimulation[10, 11]. Thus, locomotor-like patterns can be modulated by stimulation of the networks intrinsic to the spinal cord without the contribution of descending signals. To take advantage of these properties, a more thorough knowledge of the mechanisms of spinal cord stimulation, along with a more detailed understanding about specific sites and parameters of stimulation and their corresponding motor output is needed.
Ichiyama et al. reported that epidural electrical stimulation of the spinal cord can induce rhythmic, alternating hindlimb locomotor activity in chronic spinal rats. Stimulation at the L2 spinal segment at frequencies between 30 and 50 Hz consistently produced successful bilateral stepping. Similar epidural stimulation at other spinal segments were less effective, e.g., epidural stimulation at the T13 or L1 evoked rhythmic activity in only one leg and stimulation at the L3, L4, or L5 produced mainly flexion movements.
More recently, completely paralyzed (motor complete, sensory incomplete) human subjects were implanted with a commercially available spinal cord electrode array and stimulation package originally designed for pain suppression, unpublished observations. Stimulation of specific spinal segments (caudal electrodes, ~ S1 spinal level) in combination with the sensory information from the lower limbs and weeks of stand training was sufficient to generate full weight-bearing standing. These subjects also recovered some voluntary control of movements of the toe, ankle, and the entire lower limb, but only when epidural stimulation was present. Thus it appears that the epidural stimulation provided excitation of lumbosacral interneurons and motoneurons that, when combined with the weak excitatory activity of descending axons that were not otherwise detectable, achieved a level of excitation that was sufficient to activate the spinal motor circuits. These results demonstrate that some patients clinically diagnosed as having complete paralysis can use proprioceptive input combined with some synaptic input from descending motor signals, perhaps residual but functionally silent without epidural stimulation to the spinal motor circuits, to generate and control a range of motor functions during epidural stimulation.
These studies suggest that the intrinsic circuits of the spinal cord, if intact, are desirable targets for stimulus-based therapies and strategies. Secondly, the specific stimulation parameters are highly critical to the pattern and quality of functional motor output. The technological hurdles to reach these targets are non-trivial. We have designed an electrode array capable of selectively stimulating specific segments of the rat spinal cord to generate discrete motor responses using a high-density grid of epidural electrodes embedded within a thin-film flexible substrate[13, 14]. Although stimulation occurs at the surface level, miniaturization of the electrode contacts and the use of materials specific to our design restrict the effective field of stimulation to a smaller area as compared with conventional wire surface electrodes.
The specificity and high-density features of the electrode array enable us to capitalize on two key features of the spinal cord circuitries that are believed to be essential for rehabilitating posture and locomotion after spinal cord injury (SCI). Firstly, the spinal circuitry can be neuromodulated and the stimulation can be carefully delimited to affect only relevant areas of the spinal cord, thus optimizing the motor outcome. Secondly, as locomotor circuitries are highly plastic and adapt when provided with sensory cues during motor training, the density and versatility of the multi-electrode array allows for rapid adjustments of stimulation protocols and adaptations to physiological changes that may occur in the spinal cord over time after injury.
Several design features were taken into account including the flexibility of the array, biocompatibility of the base, and stability of the electrodes for a chronic implant. Parylene C has emerged as an ideal electrode array substrate due to its biocompatibility, insulative properties, flexibility, and tear resistance. The tear resistance of parylene C is large, making the arrays robust to surgical manipulation, as well as to stresses produced in a moving animal. The techniques needed to manufacture these multi-electrode devices are not unprecedented. This is the first time, however, that this technology has been adapted for the express purpose of controlling stimulation at specific sites of the spinal cord in a chronic preparation. Given these basic principles and the results observed in the animal models with conventional wire electrodes[10, 17] and from the human subjects with commercially available electrode arrays, it seems likely that use of a high-density electrode array could greatly improve the quality of standing and stepping after paralysis.
Rather than attempting to impose exogenous motor commands, this strategy will capitalize on the intrinsic neural control mechanisms of the spinal cord that remain functional post-SCI, enabling the spinal circuits to process sensory input and to serve as the primary source of control. Using this technology, we can selectively and differentially activate distinct neuronal groups distributed throughout the spinal cord, allowing stimulation of specific electrodes on the array to modulate the physiological state of the spinal circuitry so that sensory input can control various hindlimb motor outputs. To examine the potential capabilities of this stimulation system, we used this novel, flexible, high-density stimulating electrode array during the recovery of standing and stepping in adult rats after a complete mid-thoracic spinal cord transection.
Data were obtained from adult female Sprague Dawley rats (270–300 g body weight). Pre- and post-surgical animal care procedures have been described in detail previously. The rats were housed individually with food and water provided ad libitum. All survival surgical procedures were conducted under aseptic conditions and with the rats deeply anesthetized (isoflurane gas administered via facemask as needed). All procedures described below are in accordance with the National Institute of Health Guide for the Care and Use of Laboratory Animals and were approved by the Animal Research Committee at UCLA.
Five rats were implanted and tested for the biocompatibility of the implant and stability of the spinal electrodes and stable EMG responses. Once we were satisfied with the stability of the design, a stable array was implanted in one animal to collect chronic physiological data. Due to the complex nature of the fabrication, implantation, and experimentation processes, a limitation of the study is that the standing and stepping data presented in this manuscript are from one animal chronically implanted for 5 weeks. These data will be used as a stepping-stone for future experiments and design modifications.
The complete implant consists of this electrode array, a multiplexer circuit, various wires, and a headplug (Figure 1). The multiplexer circuit routes connections and performs pre-ampification to reduce the total number of headplug wires needed from 37, for a passive implant as seen in prior work by our group, to just 12 wires. This design reduces surgery complications and also serves as a stepping-stone for a fully wireless design. The electrode array is interfaced to the multiplexer board with conductive epoxy. The implant then is sealed with 20 μm of parylene, biocompatible silicone (MDX 4–4210), biocompatible epoxy (Loctite M-121HP), and another 20 μm of parylene.
Control box and multiplexer circuit board description
In the multiplexer circuit schematic (Figure 4), the black tag refers to the connection to the spinal electrode. En+, En- refers to an EMG wire pair. A3 refers to a spinal electrode in column A and row 3. G1 and G2 are reference wires (implanted on either side of the back of the animal). Three power lines are present that are used to power up the system: 12 V, 5 V, and ground (not shown in Figure 4). The desired operating mode of the circuit is configured by sending a 30-bit serial data stream (6 μs configuration time) through Clock and Data that feed into the shift registers SR1–SR4 (NXP Semiconductors 74HC164). These shift registers, in turn, configure the 10 analog multiplexer chips (M0 to M9) and EN enables them. M0 (Analog Devices ADG1209) and M1–M9 (Analog Devices ADG1209) are interconnected such that after configuration the desired electrodes or EMG wires are routed either to Stim+ and Stim- during stimulation or to pre-amplifiers AMP1-AMP4 (Analog devices AD8224) during recording. The pre-amplifiers are differential instrumentation amplifiers set to a gain of 200 and send outputs to A1–A4. The circuit board uses four copper layers and measures 10.3 mm by 33.2 mm.
Head connector and intramuscular EMG electrode implantation
A small incision was made at the midline of the skull. The muscles and fascia were retracted laterally, small grooves were made in the skull with a scalpel, and the skull was dried thoroughly. Two amphenol head connectors with Teflon-coated stainless steel wires (AS632, Cooner Wire, Chatsworth CA) were securely attached to the skull with screws and dental cement as described previously[12, 18]. The medial gastrocnemius (MG), tibialis anterior (TA), and soleus (Sol) muscles were implanted bilaterally with EMG recording electrodes as described by Roy et al.. Skin and fascial incisions were made to expose the belly of each muscle. Two wires extending from the multiplexer circuit board (Figure 1) were routed subcutaneously to each muscle. The wires were inserted into the muscle belly using a 23-gauge needle and a small notch (~0.5–1.0 mm) was removed from the insulation of each wire to expose the conductor and form the electrodes. The wires were secured in the belly of the muscle via a suture on the wire at its entrance into and exit from the muscle belly. The wires were looped at the entrance site to provide stress relief. The proper placement of the electrodes was verified 1) during the surgery by stimulating through the stimulator in the control box (Figures 1,2,3 and4) and by selecting the correct channels on the multiplexer circuit board and 2) post-surgery by dissection.
Spinal cord transection and array implantation
A partial laminectomy was performed at the T8-T9 vertebral level and a complete spinal cord transection to include the dura was performed at ~T8 spinal level using microscissors. Two surgeons verified the completeness of the transection by lifting the cut ends of the spinal cord and passing a glass probe through the lesion site. Gel foam was inserted into the gap created by the transection as a coagulant and to separate the cut ends of the spinal cord.
All incision areas were irrigated liberally with warm, sterile saline. All surgical sites were closed in layers, i.e., muscle and connective tissue layers with 5.0 Vicryl (Ethicon, New Brunswick, NJ) and the skin incisions on the back and the limbs with 5.0 Ethilon. All closed incision sites were cleansed thoroughly with warm saline solution. Analgesia was provided by buprenex (0.5–1.0 mg/kg, 3 times/day s.c.). The analgesics were initiated before the completion of the surgery and continued for a minimum of 2 days post-surgery. The rats were allowed to fully recover from anesthesia in an incubator. The spinal rats were housed individually in cages that had ample CareFresh bedding and their bladders were expressed manually 3 times/day for the first 2 weeks after surgery and 2 times per day thereafter. The hindlimbs of the spinal rats were moved passively through a full range of motion once per day to maintain joint mobility.
Stimulation and testing procedures
Two stimulation protocols were used for testing (Figures 3 &4). A monopolar configuration where the cathode was chosen from one of the 27 electrodes on the array and the anode placed subcutaneously on the side of the body (ground wire, Figures 1,2,3 and4). On the testing day, the cathode was selected sequentially among all electrodes on the array to systematically cover the entire surface of the array and was used to record evoked potentials from the MG, TA, and Sol muscles bilterally. Evoked potentials were recorded from the muscles implanted with EMG electrodes by stimulating the spinal cord at a low frequency (1 Hz) and voltage sweep from 1–8 V (1 V increments) with the rat suspended in a jacket with its hindpaws in contact with a stationary treadmill (bipedal standing position). A bipolar configuration where both the cathode and anode were selected from the set of 27 electrodes on the array was used to facilitate the standing and stepping ability of the spinal rats. Sub–sets of bipolar configurations were tested on different test days. For both the bipolar configurations, the stimulation frequency was based on previously reported values[7, 10, 12, 20, 21] and the stimulation intensity was varied (range from 1–8 V) to optimize the standing and stepping ability of the spinal rats. EMG was recorded from the MG, TA, and Sol bilaterally while the rats stepped bipedally on a specially designed motor-driven rodent treadmill at 13.5 cm/s. The treadmill belt had an anti-slip material that minimized slipping while stepping. The rats were placed in a body weight support system that allowed the rat to support the maximum amount of its body weight while stepping with plantar placement.
Data collection and analysis
EMG recordings from the hindlimb muscles were pre-amplified by the multiplexer circuit board and an external control box before being sent to a band-pass filter (1 Hz to 5 KHz), externally amplified (A-M Systems Model 1700 differential AC amplifier: A-M Systems, Carlsborg, WA), and sampled at a frequency of 10 KHz using a custom data acquisition program written in the LabView development environment (National Instruments, Austin, TX) as described previously. Evoked potentials during standing with low frequency stimulation (1 Hz) were analyzed as described previously[7, 10]. The responses were divided into 20 ms windows using the stimulation pulse as the trigger. These windows were averaged over 10 evoked responses and the peak response was detected using custom MATLAB code. These peaks then were binned into early (ER, 1–3 ms latency), middle (MR, 4–6 ms latency), and late (LR, 7–10 ms) responses. The mean amplitudes and latencies for the ER, MR, and LR for both the MG and TA at different intensities of stimulation for each electrode on the array were determined. The EMG signals during weight-bearing standing under epidural stimulation at higher frequencies were analyzed using a custom script written in MATLAB to estimate the MR (latency 4–6 ms) and LR (latency 10–25 ms). The raw EMG signals during bipedal stepping on the treadmill were rectified and then sent through a low pass filter to form a linear envelope to assess the stepping patterns as previously described.
A 400 mV sinusoidal wave (10 KHz with a 10 KΩ resistor in series with the spinal electrode and the indifferent ground) was used to test electrode impedance. The voltage across the electrode on the spinal cord and the ground placed subcutaneously in the back region was used to measure the electrode impedance. The electrode impedance was inversely related to the ability of the electrode to stimulate the spinal cord.
Facilitation of standing with epidural stimulation
Additional file 1: Video 1: This video file demonstrates a spinal rat transitioning from a crouched to a standing posture when facilitated by epidural stimulation (A1C5: Freq – 40 Hz, Amplitude – 3 V, Pulse Width – 0.2ms). (AVI 2 MB)
In contrast to stimulation of rostral electrode pairs, bipolar stimulation of caudal electrode pairs at any frequency failed to facilitate weight-bearing standing. This difference between stimulation of rostral vs. caudal electrode pairs clearly demonstrates the importance of the location of the electrodes and the frequency of stimulation in tuning the neural circuits to generate a specific motor response.
Facilitation of stepping via epidural stimulation
Additional file 2: Video 2: This video file demonstrates a spinal rat stepping at 13.5 cm/s on a treadmill with good coordination and body weight support when facilitated by epidural stimulation(A1C7: Freq – 40 Hz, Amplitude – 3.2 V, Pulse Width – 0.2 ms). (AVI 3 MB)
Additional file 3: Video 3: This video file demonstrates a spinal rat stepping at 13.5 cm/s on a treadmill with minimal body weight support when facilitated by epidural stimulation(A5A6: Freq – 40 Hz, Amplitude – 3.3 V, Pulse Width – 0.2 ms). (AVI 1 MB)
Combined, these results highlight the importance of the position of the cathode and anode on the spinal cord to facilitate stepping after injury and that the ability to choose between specific sites of stimulation is critical for modulating the types of motor output produced by the epidural stimulation.
Differential modulation of evoked potentials to low frequency epidural stimulation during standing based on electrode position and stimulation intensity
Similar to the ER, the MR begins around rows 4 and 5 and generally increases in amplitude with increasing stimulation intensity (Figure 10). Unlike the ER, however, the latency of the MR in the TA remains constant from rows 4 to 9 across the stimulation intensities and the latencies in the MG decrease in the more caudal electrodes for any given intensity of stimulation. The MR from stimulation of the most rostral sets of electrodes (rows 1, 2, and 3) shows a much higher latency (~7 ms) compared to the MR from rows 4, 5, and 6 (4–6 ms), i.e., the start of the motor pools of the ankle flexors and extensors, suggesting that there could be an additional synapse for the evoked potential from the rostral region of the spinal cord before the signal reaches the muscles (Figure 10 – e.g., RMG 4 V). The MR in the muscles are higher in the right limb in both the TA and MG at any given voltage (Figure 10 – e.g.,TA and MG at 5 V) through the C, or right side, column of electrodes located ipsilateral to the muscles. Thus, it appears that these evoked responses may be highly dependent on the spatial location of the stimulation. These results highlight the importance of the ability to stimulate specific sites from a therapeutic and device standpoint.
The LR are very general with no real pattern in the observed latencies or amplitudes, suggesting that the LR is a result of activation of various spinal interneuronal circuits that eventually filter down to the muscles. Stimulation at all intensities generates an LR at all electrodes. Several investigators have shown the importance of the presence of an LR to the stepping ability in spinal rats[6–8]. While the specific interaction of the interneurons and the possible structure of these network circuitries are beyond the scope of this paper, it is nonetheless important to identify the diversity of the signals evoked at this level. These results provide important insight into the highly crucial nature of the finite spatial resolution of the stimuli. In addition, the above data indicate that the LR is far less electrode specific than the ER and MR. The functional significance of these observations needs further study.
Biocompatibility and durability of the chronic multi-electrode array
Average impedances for each electrode in chronically implanted arrays
We have demonstrated a novel technique, using a high-density parylene-based multi-electrode platinum array, to selectively activate spinal neurons to facilitate standing and stepping in rats after a complete spinal cord transection at a low-thoracic level. The results demonstrate that spinal rats can stand and step when the spinal cord is stimulated tonically at 40 Hz by electrodes located at specific sites on the spinal cord. The quality of stepping and standing was dependent on the location of the electrodes on the spinal cord, the specific stimulation parameters, and the orientation of the cathode and anode. In addition, the amplitude and latency of evoked potentials were determined in non-anesthetized spinal rats during standing to assess the efficacy of selected spinal circuits. The evoked potentials are critical tools to study selective activation of interneuronal circuits via responses of varying latencies.
Critical features of the stimulation parameters for facilitating standing and stepping
Based on the results, we can generalize that combinations of stimulation with the cathode at the rostral end of the spinal cord results in better stepping ability as compared to combinations with the cathode at the caudal electrodes. This suggests that neurons and neuronal circuits at the rostral end of the spinal cord respond more effectively to the cathode as compared to the caudal sets of electrodes that respond more effectively to the anode. The best results were observed with the cathode and anode located in different rows of the electrode array and the cathode and anode in different columns of the electrode array. The present data also suggest that the more effective standing and stepping can be obtained with bipolar compared to monopolar stimulation. This issue, however, needs to be examined more thoroughly. While the specific composition of the neuronal circuitry and aggregate networks of the spinal cord must be studied further, it is clear that modulating the stimulation protocol and targeting specific anatomical sites of the spinal cord lead to variable motor outputs distinct from one another with unique functional effects.
Modulation of specific motor pools using the multi-electrode array
The evoked potentials from specific muscles during monopolar stimulation at different intensities allowed us to assess the activation of the motor pools of the ankle flexor and extensors in the spinal cord[24, 25]. Evoked potentials from monopolar stimulation reflect the activation of specific neuronal circuits as demonstrated by the responses shown in Figures 9,10, and11. Additionally, the higher amplitudes of the MR on the ipsilateral compared to the contralateral side demonstrate the ability to selectively activate different circuitries and to stimulate specific anatomical areas and combinations of motor pools. Different levels of inhibition vs. excitation of spinal circuitries also could be induced selectively. This potential to selectively activate specific combinations of motor pools and levels of inhibition and excitation translates into the unique capability of electrode arrays to control motor behavior.
Importance of the multiplexer for chronic implantation with wireless capability in small animals
When the durability of an implant is a requirement, the size and biocompatibility of the device are crucial factors in successfully collecting data. Our animal experiments currently rely upon wire bundles to connect the electrode arrays to external computers and electronics. As the number of required connections and the complexity of the device increases, the size of the wire bundle increases as well, reducing the probability of success of the implant due to potential tissue damage and infections caused by the wire bundles. We have partially addressed this problem in our original design by employing a multiplexer (Figure 4) to reduce the number of required connections and changing the form factor of the electrode package into a more easily implantable design. We now plan to develop a relatively generic implantable wireless multi-channel stimulating/recording engine that can be scaled to different species, e.g., rat, cat, or human. This will make the electrode array more useful in a number of ways. For example, the elimination of the wire bundle will increase the biocompatibility of the implant and reduce chances of infection and tissue damage. Additionally, because the wireless system will be a general device designed to have a variety of applications, the transition from animal to human studies will likely be simplified since the fundamental basis of the device will remain consistent.
Early recovery of stepping and standing after SCI facilitated by epidural stimulation
Several studies have shown that epidural stimulation at L2 and/or S1 using wire electrodes in combination with motor training can facilitate stepping within 3–4 weeks after a complete spinal cord transection[7, 9, 10, 21]. Using the parylene-based platinum electrode arrays described herein we have been successful in facilitating weight-bearing standing and stepping within 8–10 days post-transection. Thus use of the electrode array allows us to tap into the spinal networks to enable stepping sooner after injury as compared to using conventional wire electrodes. Future directions to improve this technology will be to 1) develop computational and mathematical means to detect patterns, determine relationships using evoked potentials, and predict functional outputs, 2) record spinal-evoked potentials during stepping, and 3) combine pharmacological interventions with multi-electrode epidural stimulation as a therapeutic rehabilitation strategy.
Mathematical modeling to characterize motor responses using learning algorithms
The development of mathematical and computational infrastructures to better characterize motor outputs of stimulation will be crucial to the further development of this neuromodulatory technology. The sheer numbers of involved electrodes, the wide range of stimulation parameters, and the number of functional outcome measures represent a matrix of inputs and outputs that creates a bottleneck to accurately analyze all results. Therefore, it will become necessary to develop tools such as machine-learning algorithms and classification schemes to automate the processing. This is not only important from the perspective of experimental efficiency or basic scientific goals, but particularly from the point of transitioning this technology to clinical therapeutic paradigms. Using a highly differentiated electrode array, it becomes crucial to determine the holistic differences between the smallest variations in the stimulation properties and locations to modulate the networks and produce ordered, desired behavioral outputs. To achieve this, we must develop the means to process and interpret the voluminous information recorded from high-density electrode arrays.
Need for the ability to record evoked potentials from the spinal cord
The full potential for the use of high-density epidural electrode arrays in clinical and basic scientific studies cannot yet be realized due to limitations in currently available implantable stimulating electronics. The stimulators currently FDA-approved for human studies are too limited in the types of stimulation that they can generate and have no capability to record evoked potentials. Currently, we are unable to detect dynamic changes in intra-spinal cord network interactions during stimulation. The importance of the afferent information to motor command and control cannot be overestimated, yet we have little to no information about the ascending signals that form a significant component of the CPG’s input data. Adding the ability to record from intrinsic networks of the spinal cord could reveal a great deal about the feedback mechanisms that form the foundation for locomotor pattern generation. This will require that the technology for the electrodes be refined to provide optimal characteristics for both stimulation and recording.
Potential for neuromodulation of the spinal cord and facilitation of specific responses using pharmacological interventions combined with the electrode array
An important aspect of facilitating stepping after SCI is the administration of pharmacological interventions. Although the pharmacological effects are transient, concurrent application of other treatments seems to supplement pharmacologically induced activity[17, 20]. These pharmacological treatments appear to raise the excitability of the spinal locomotor circuits by lowering their threshold for activation, and thereby facilitating the effects of multi-electrode epidural stimulation. Specific activation of neuronal networks through the use of an electrode array after administration of pharmacological interventions will allow us to selectively activate specific motor pools for the control of fine movements as well as stepping patterns. Examination of these altered physiological states have the potential to reveal more information about the underlying circuitry of the spinal cord by further delimiting the inhibitory and excitatory components of the circuits responsible for motor behavior, ultimately allowing for the identification and characterization of the neuronal populations responsible for the recruitment of specific motor pools.
Neurophysiological mechanisms and specific sensorimotor integration impacting motor function via the electrode array after SCI
Given the range of motor behaviors that can be generated with modest levels of stimulation, i.e., primarily sub-motor threshold levels, of different combinations of electrodes and at different frequencies, it is evident that the threshold for excitation of different spinal interneuronal networks are being modulated. Conceptually our strategy for facilitating these motor behaviors is to achieve a physiological state that enables the proprioceptive input derived from stepping and standing to serve as the source of control. That is, the "sub-threshold" intensity of stimulation that modulates the spinal circuitry associated with stepping and standing may not, and actually preferably does not, induce action potentials among the pathways extending from sensory afferents to all of the motor pools. Thus, rather than imposing a specific motor response by stimulating at high intensities, and thus precluding proprioceptive modulation, the activated pathways are determined by the ensemble of sensory information being projected in real time to the spinal circuitry. Regarding the degree of selectivity of specific pathways that could be modulated, it is important to recognize that the extensive divergence of a single Ia fiber from each muscle spindle has extensive synaptic connectivity to not only the homonymous motor pools, but also to synergists and indirectly to antagonistic motor pools through Ia inhibitory interneurons. In addition, robust intersegmental connectivity among the lumbar segments via ascending projections from the sacral segments has recently been reported. Combined, these observations are consistent with the interpretation that epidural stimulation combined with pharmacological modulation is impacting many different pathways simultaneously but in different degrees and proportions based on the stimulation parameters described in the present paper.
The high density electrode array described in this paper 1) allows high spatial resolution and the ability to selectively activate different neural pathways within the lumbosacral region of the spinal cord to facilitate standing and stepping in adult spinal rats, and 2) provides the capability to evoke motor potentials and thus a means for assessing connectivity between sensory circuits and specific motor pools and muscles. Our initial data underscore the importance of electrode location and anode–cathode orientation and stimulation properties, especially with respect to future therapeutic devices and modulatory “tuning” of epidural stimulation patterns, to provide optimal stimulation for motor function restoration after SCI in animals and humans. Further revisions and additions to this system, including wireless transmission of data, greater software control of the stimulation properties, and increasingly sophisticated data analysis techniques will allow us to further our work/results and gain insights into the neural circuits responsible for specific functional motor responses.
We would like to thank Maynor Herrera for providing excellent animal care and Sharon Zdunowski for technical assistance. This research was supported by the National Institute of Biomedical Imaging and Bioengineering R01EB007615.
- Grillner S: Locomotion in vertebrates: central mechanisms and reflex interaction. Physiol Rev 1975, 55: 247-304. 10.1104/pp.55.2.247PubMedGoogle Scholar
- Edgerton VR, Tilakaratne N, Bigbee A, de Leon R, Roy RR: Plasticity of the spinal neural circuitry after injury. Annu Rev Neurosci 2004, 27: 145-167. 10.1146/annurev.neuro.27.070203.144308PubMedView ArticleGoogle Scholar
- Hodgson JA, Roy RR, de Leon R, Dobkin B, Edgerton VR: Can the mammalian lumbar spinal cord learn a motor task? Med Sci Sports Exercise 1994, 26: 1491-1497.View ArticleGoogle Scholar
- Forssberg H: Stumbling corrective reaction: a phase-dependent compensatory re- action during locomotion. J Neurophysiol 1979, 42: 936-953.PubMedGoogle Scholar
- Harkema SJ, Hurley SL, Patel UK, Reguejo PS, Dobkin BH, Edgerton VR: Human lumbosacral spinal cord interprets loading during stepping. J Neurophysiol 1997, 77: 797-811.PubMedGoogle Scholar
- Musienko PE, Bogacheva IN, Gerasimenko YP: Significance of peripheral feed-back in the generation of stepping movements during epidural stimulation of the spinal cord. Neurosci Behav Physiol 2007, 37: 180-191.View ArticleGoogle Scholar
- Lavrov I, Courtine G, Dy CJ, van den Brand R, Fong AJ, Gerasimenko YP, Zhong H, Roy RR, Edgerton VR: Facilitation of stepping with epidural stimulation in spinal rats: role of sensory input. J Neurosci 2008, 28: 7774-7780. 10.1523/JNEUROSCI.1069-08.2008PubMedPubMed CentralView ArticleGoogle Scholar
- Harkema SJ, Gerasimenko YP, Hodes J, Burdick J, Angeli C, Chen Y, Ferreira C, Willhite A, Rejc E, Grossman R, Edgerton VR: Effect of epidural stimulation of the lumbosacral spinal cord on voluntary movement, standing and assisted stepping after motor complete paraplegia: a case study. Lancet 2011, 377: 1938-1947. 10.1016/S0140-6736(11)60547-3PubMedPubMed CentralView ArticleGoogle Scholar
- Musienko P, Courtine G, Tibbs JE, Kilimnik V, Savochin A, Roy RR, Edgerton VR, Gerasimenko Y: Somatosensory control of balance during locomotion in decerebrated cat. J Neurophysiol in pressGoogle Scholar
- Gerasimenko YP, Avelev VD, Nikitin OA, Lavrov IA: Initiation of locomotor activity in spinal cats by epidural stimulation of the spinal cord. Neurosci Behav Physiol 2003, 33: 247-254. 10.1023/A:1022199214515PubMedView ArticleGoogle Scholar
- Gerasimenko YP, Ichiyama RM, Lavrov IA, Courtine G, Cai L, Zhong H, Roy RR, Edgerton VR: Epidural spinal cord stimulation plus quipazine administration enable stepping in complete spinal adult rats. J Neurophysiol 2007, 98: 2525-2536. 10.1152/jn.00836.2007PubMedView ArticleGoogle Scholar
- Ichiyama RM, Gerasimenko YP, Zhong H, Roy RR, Edgerton VR: Hindlimb stepping movements in complete spinal rats induced by epidural spinal cord stimulation. Neurosci Lett 2005, 383: 339-344. 10.1016/j.neulet.2005.04.049PubMedView ArticleGoogle Scholar
- Nandra MS, Lavrov IA, Edgerton VR, Tai YC: A Parylene-based microelectrode array implant for spinal cord stimulation in rats. In Proceedings of the 24th IEEE Conference Engineering in Medicine and Biological Society. : ; 2011:1007-1010. Jan 23Google Scholar
- Rodger DC, Tai YC: Microelectronic packaging for retinal prostheses. IEEE Eng Med Biol Mag 2005, 24: 52-57.PubMedView ArticleGoogle Scholar
- Wolgemuth L: Assessing the performance and suitability of parylene coating. Med Device Diagn Ind 2000, 22: 42-49.Google Scholar
- Rodger DC, Weiland JD, Humayun MS, Tai YC: Scalable high lead-count parylene package for retinal prostheses. Sensor Actuator B Chem 2006, 117: 107-114. 10.1016/j.snb.2005.11.010View ArticleGoogle Scholar
- Ichiyama RM, Courtine G, Gerasimenko YP, Yang GJ, van den Brand R, Lavrov IA, Zhong H, Roy RR, Edgerton VR: Step training reinforces specific spinal locomotor circuitry in adult spinal rats. J Neurosci 2008, 28: 7370-7375. 10.1523/JNEUROSCI.1881-08.2008PubMedView ArticleGoogle Scholar
- Roy RR, Hodgson JA, Lauretz SD, Pierotti DJ, Gayek RJ, Edgerton VR: Chronic spinal cord-injured cats: surgical procedures and management. Lab Anim Sci 1992, 42: 335-343.PubMedGoogle Scholar
- Roy RR, Hutchison DL, Pierotti DJ, Hodgson JA, Edgerton VR: EMG patterns of rat ankle extensors and flexors during treadmill locomotion and swimming. J Appl Physiolology 1991, 70: 2522-2529.Google Scholar
- Courtine G, Gerasimenko YP, van den Brand R, Yew A, Musienko P, Zhong H, Song B, Ao Y, Ichiyama RM, Lavrov IA, Roy RR, Sofroniew MV, Edgerton VR: Transformation of nonfunctional spinal circuits into functional states after the loss of brain input. Nat Neurosci 2009, 12: 1333-1342. 10.1038/nn.2401PubMedPubMed CentralView ArticleGoogle Scholar
- Lavrov I, Gerasimenko YP, Ichiyama RM, Courtine G, Zhong H, Roy RR, Edgerton VR: Plasticity of spinal cord reflexes after a complete transection in adult rats: relationship to stepping ability. J Neurophysiol 2006, 96: 1699-1710. 10.1152/jn.00325.2006PubMedView ArticleGoogle Scholar
- de Leon RD, Reinkensmeyer DJ, Timoszyk WK, London NJ, Roy RR, Edgerton VR: Use of robotics in assessing the adaptive capacity of the rat lumbar spinal cord. Prog Brain Res 2002, 137: 141-149.PubMedView ArticleGoogle Scholar
- de Guzman CP, Roy RR, Hodgson JZ, Edgerton VR: Coordination of motor pools controlling the ankle musculature in adult spinal cats during treadmill walking. Brain Res 1991, 555: 202-214. 10.1016/0006-8993(91)90343-TPubMedView ArticleGoogle Scholar
- Manzano G, McComas AJ: Longitudinal structure and innervation of two mammalian hindlimb muscles. Muscle Nerve 1988, 11: 1115-1122. 10.1002/mus.880111103PubMedView ArticleGoogle Scholar
- Rivero-Melián C: Organization of hindlimb nerve projections to the rat spinal cord: a choleragenoid horseradish peroxidase study. J Comp Neurol 1996, 364: 651-663. 10.1002/(SICI)1096-9861(19960122)364:4<651::AID-CNE4>3.0.CO;2-2PubMedView ArticleGoogle Scholar
- Nelson S, Mendell L: Projection of single knee flexor Ia fibers to homonymous and heteronymous motoneurons. J Neurophysiol 1978,41(3):778-787.PubMedGoogle Scholar
- Etlin A, Blivis D, Ben-Zwi M, Lev-Tov A: Long and short multifunicular projections of sacral neurons are activated by sensory input to produce locomotor activity in the absence of supraspinal control. J Neurosci 2010,30(31):10324-10336. 10.1523/JNEUROSCI.1208-10.2010PubMedView ArticleGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.